arXiv:2212.05461v1 [physics.ins-det] 11 Dec 2022

Detector array for the^7H nucleus multi-neutron decay study Массив детекторов для исследования многонейтронного распада ядра^7H

A. A. Bezbakh^{a,b}, S. G. Belogurov^a, V. Chudoba^{a,b}, A. S. Fomichev^a, A. V. Gorshkov^a, L. V. Grigorenko^a, G. Kaminski^{a,c}, M. S. Khirk^a A. G. Knyazev^a, S. A. Krupko^a, B. Mauyey^{a,d}, I. A. Muzalevskii^{a,b}, E. Yu. Nikolskii^a, A. M. Quynh^{a,e}, P. G. Sharov^{a,b}, R. S. Slepnev^a, S. V. Stepantsov^a, G. M. Ter-Akopian^a, and R. Wolski^a А. А. Безбах^{a,b}, С. Г. Белогуров^a, В. Худоба^{a,b}, А. С. Фомичев^a, А. В. Горшков^a, Л. В. Григоренко^a, Г. Каминьски^{a,c}, М. С. Хирк^a, А. Г. Князев^a, С. А. Крупко^a, Б. Мауей^{a,d}, И. А. Музалевский^{a,b}, Е. Ю. Никольский^a, А. М. Куинь^{a,e}, П. Г. Шаров^{a,b}, Р. С. Слепнев^a, С. В. Степанцов^a, Г. М. Тер-Акопьян^a, Р. Вольски^a

a Flerov Laboratory of Nuclear Reactions, JINR, 141980 Dubna, Russia a Лаборатория ядерных реакций им. Г.Н. Флерова, ОИЯИ, 141980 Дубна, Россия

b Institute of Physics, Silesian University in Opava, 74601 Opava, Czech Republic b Физический институт, Силезский университет в Опаве, 74601 Опава, Чехия

c Heavy Ion Laboratory, University of Warsaw, 02-093 Warsaw, Poland c Лаборатория тяжелых ионов, Варшавский университет, 02-093 Варшава, Польша

d L.N. Gumilyov Eurasian National University, 010008 Astana, Kazakhstan d Евразийский национальный университет имени Л. Н. Гумилёва, 010008 Астана, Казахстан

e Nuclear Research Institute, 670000 Dalat, Vietnam e Институт ядерных исследований, 670000 Далат, Вьетнам

На пучке радиоактивных ядер сепаратора ACCULINNA-2 Лаборатории ядерных реакций им. Г. Н. Флёрова создаётся установка, предназначенная для детального изучения пятительного распада ядер^7, образующихся в результате реакции передачи протона от налетающего ядра^8He на ядро мишени 2. В настоящей статье даётся описание устройства из 100 пластических сцинтилляторов ВС-404, предназначенного для регистрации нейтронов, телескопа Si детекторов ядер отдачи^3He, а также комплекта детекторов для ΔE-E- TOF регистрации ядер^3H, вылетающих из мишени при распаде^7H. Показаны результаты моделирования методом Монте-Карло, давшего ожидаемые значения энергии и параметры траекторий всех частиц. Приведены результаты полученных оценок светимости обсуждаемых экспериментов.

Setup fitting the requirements for the detailed study of the five-body decay of the 7H nucleus obtained as a result of the proton transfer from the^8He projectiles to the deuterium target nuclei is being built at the radioactive beam line of ACCULINNA- 2 separator in the G.N. Flerov Laboratory of Nuclear Reactions. Described here is the assembly of 100 BC-404 plastic scintillators, intended for neutron detection, the annular Si detector telescope for the isotope[3]He recoils, and the detector array providing the ΔE-E-TOF registration of^3H nuclei emitted at the^7H decay. Results obtained by the Monte Carlo simulations made for the energy values and flight passes of all these particles are given together with the luminosity expected for the discussed experiments.

PACS: 25.45.−z; 25.70.−z; 27.20.+n; 29.40.Gx; 29.40.Mc; 29.40.Wk

Submitted to PEPAN letters.

1. Setup with the 75 mm thick BC-404 scintillators

Presented is the modification of the detector array intended for the study of nuclei undergoing decay with the multi-neutron emission. The modification assumes the use of the assembly of 100 BC-404 scintillators which are at the disposal of ACCULINNA group. The front view of a single scintillator is shown in Figure 1. In its cross-section, the 75 mm thick plastic is the hexagon inscribed in the circle with diameter 100 mm and it is enclosed in some 1-mm thick cover. Consequently, 95.5% area of a compact assembly made with the use of these plastics will be covered by the BC-404 scintillators.

The Neutron Wall (NW) arrangement, shown in Figure 1, will be the central part of the setup being created to satisfy the requirements becoming apparent as a result of study made on the decay spectra of^7H and^6H nuclei [1–3]. The setup composition is given in Figure 2.

Assembly of 100 plastic scintillators shown in Figure 1 covers a surface approximated by a circle with diameter 93 cm. Its empty central part opens way for the^8He nuclei to fly free to the beam stopper placed at a 0.5-meter distance behind the NW without making background distorting the detection of multiple-neutron events.

Detection probabilityε_n ≈ 0.28 is estimated for single neutrons with energy 20–30 MeV hitting the array. This estimate is based on the known [4, 5] cross section values of the elastic^1H(n, p) scattering and^{12}C(n, n^′3α), 12C(n, α)^9Be,^{12}C(n, np)^{11}B,^{12}C(n, p)^{12}B,^{12}C(n, nγ) reactions induced by neutrons bombarding the hydrogen and carbon nuclei making the BC-404

Figure 1. Left: The 75-mm thick plastic scintillator having the hexagonal crosssection. Painted in yellow is shown the plastic cross-section enclosed in the lightproof (aluminum) shield. Right: Assembly of 100 BC-404 plastic scintillators suggested for the detection of nuclear decay events with multiple neutron emission.

Table 1. Depletion constant and flux-decrease factors in the material of plastic BC- 404 obtained for the neutrons at five energy values. F is factor of flux reduction at plastic depth 75 mm.

E_n, MeV 10 20 25 30 35

λ, mm^{−1} 0.00760 0.00770 0.00738 0.00724 0.00630 F 0.56 0.56 0.57 0.58 0.68

plastic composition in amounts 5.23 × 10^{22} cm^{−3} and 4.74 × 10^{22} cm^{−3}, respectively.

The depletion rate of neutron flux in the BC-404 plastic is calculated by the use of the depletion constant λ. The value of this constant is obtained as the sum of contributions made to the decrease of the neutron flux due to the neutron elastic scattering on the hydrogen nuclei and neutron reactions with carbon nuclei giving rise to the charged-particle emission. The values of this depletion constant obtained for neutrons with five different energy values are given in Table 1.

The values of flux reduction factor obtained as F = e^{−λ·75} for neutrons passing through the 75-mm thick BC-404 scintillator are given in Table 1 in last raw. Thus, the probability that neutrons with energy 20–30 MeV could produce any of the specified above reaction is estimated as 1 − F ≈ 0.43. Detection probabilityε_napprox0.28 was obtained comparing these reaction probabilities with those which were typical for the neutron modular detector DEMON [6].

Figure 2. Setup for the study of the^7H decay events: 1 —^2H cryogenic deuterium target; q 2a — 0.140 mm thick annular Si-detector backed with the 1 mm thick annular Si detector; 2b — 1.5 mm thick annular Si-detector; 3 — 0.18 mm thick stainless-steel vacuum window of the ACCULINNA-2 reaction chamber; 4 — plastic-scintillator array stopping the^7H-decay tritons; 5 – Neutron Wall (NW); 6 – beam stopper. Neutrons emitted from the target within angular range θ_{lab} ≤ 11.5^∘ at the^7H decay hit the NW. The distances from the target to the items enumerated in this Figure are given in millimeters on the right side. Shown on the right, in insert, is the front view of array 4 with the NW installed behind.

2. Triggering array

The suggested choice for the detector array, giving trigger signals produced by the^3He recoil nuclei emitted in the^2H(^8He,^3He)^7H reaction, is to place at position 2a (see Figure 2) a detector telescope made of a pair of annular Si detectors. Each detector is segmented in 64 rings and 64 sectors and has the 28-mm central hole and sensitive area with the 32 mm inner diameter and 125 mm outer diameter. The 0.140 mm thick front detector is the source of trigger signals generated from the recoil^3He nuclei. The second detector will be 1.0 mm thick.

Energy range 9–22 MeV is inherent to the^3He recoils emitted in the 2H(^8He,^3He)^7H reaction populating the^7H nucleus excitation spectrum in a range of 0–9 MeV above the^3H+4n decay threshold. Such recoils with energy > 14 MeV pass through the 0.140 mm front Si detector and are stopped in the second, 1 mm thick Si detector. These nuclei will be identified by the ΔE − E method against the background made by the other recoil nuclei (^2H, 3H,^4He,^6He, etc.) which are emitted from the target. The 9–14 MeV^3He recoils, stopped in the front Si detector, will be safely discriminated from the 3H recoils, emitted from the target with similar energy, because for these^3H recoil nuclei the ΔE − E identification signals are provided by the detector telescope placed in position 2a.

As for the^3He and^4He recoil nuclei emitted from the target with energy E3_{He} < 14 MeV and E4_{He} < 16 MeV, respectively, their identification will be performed on the basis of the E-ToF method consisting in the comparison of the measured energy deposit made by the recoil nucleus in the front, 0.140 mm detector, and its flight time (ToF) expended on the 155 mm distance from the target to this detector. The “start” (zero-mark) time is given by the radioactive-beam (RIB) diagnostic array situated upstream the target, and the signal coming from the front, 0.14 mm ΔE detector will provide the stop mark for the ToF evaluation. The zero-time start signal is provided by the ACCULINNA-2 RIB diagnostic with a root-mean-square error 0.2 ns, and the time measurement made with the ΔE Si detector gives the 0.5 ns stop-signal time spread. This will result in the safe separation of individual nuclides between the whole group of the^3He,^4He,^6He,^8He,^6Li, and^7Li recoil nuclei hitting the 0.14 mm Si detector. Results obtained by means of Monte Carlo (MC) simulations testify in behalf of this estimation. All simulations assumed that the^8He beam, having emittance 75 π mm · mrad and energy spread ΔE/E = 5% (FWHM), was focused on a target within a circle with diameter 15 mm.

One simulation result is presented in Figure 3. Shown are the Energy – ToF distributions simulated for the^3He recoil nuclei and^4He background nuclei emitted from the target and detected by the 0.140 mm thick annular Si detector 2a shown in Figure 2. The source of the^3He recoil nuclei is the^2H(^8He,^3He)^7H reaction populating the^7H 2.2 MeV ground state in the center-of-mass angular range 0–20 degrees. These nuclei are assumed to be formed in the thick, 2.5 mg/cm^2 cryogenic deuterium target. It is evident

Figure 3. The ToF spectra simulated for the^3He and^4He nuclei emitted from the target and stopped in the 0.140 mm annular Si detector. Further explanations are given in text.

that the distributions, simulated for the two recoils, only weakly overlap in this two-dimension plot. More than 95% of these^3He recoil nuclei can be separated without a noticeable^4He impurity. Being taken in 1.0-MeV^3He energy bands, these distributions have roughly 0.5 ns width (FWHM).

3. Information deduced on the^7H spectrum using the combined mass method

We will discuss now the application of the combined-mass method, originally developed in [7], to the study of^7H spectrum obtained in the^2H(^8He,^3He)^7H reaction. Discussed here is the experiment which will be carried out with the 25 AMeV radioactive^8He beam bombarding a 2.5 mg/cm^2 (7.5 × 10^{20} cm^{−2}) deuterium gas target. So thick target leads to the^7H missing mass determination practically useless with very bad energy resolution. However, the poorly measured energy and emission angle of the^3He recoil nucleus cause the good determination of the center-of-mass momentum value and the trajectory angle made for complete pattern of five decay products (^3H + 4n) emitted by the^7H nucleus in the^2H(^8He,^3He)^7H reaction. Precision achieved by this procedure depends on the errors inherent to the measured^3He energy (E_{lab}) and angle (θ_{lab}).

Realistic estimates come out for these errors from the complete MC simulation made for the^2H(^8He,^3He)^7H reaction populating the known^7H resonance states [2] with energies 2.2 and 5.5 MeV above the^3H + 4n decay

Figure 4. The θ_{lab} vs. E_{lab} distributions obtained by the complete MC simulations made for^3He recoil nuclei appearing in the^2H(^8He,^3He)^7H reaction resulting in the population of the^7H resonance states with energy 2.2 and 5.5 MeV. Simulations were made for the cryogenic D_2 target with a thickness of 2.5 mg · cm^{−2}, in the center-of-mass angular range 0–20 degrees.

threshold. The θ_{lab} vs. E_{lab} distributions obtained for^3He recoils showing the population of these^7H states are presented in Figure 4. The pattern displayed in Figure 4 shows that less than 1.5% of 5.5-MeV states populated in the^2H(^8He,^3He)^7H reaction mix with the 2.2-MeV states. With θ_{lab} measured with a 20 mrad precision, one knows, with precision better than 4 mrad, the center-of-mass emission angle of the produced^7H nucleus. At that time, the^7H momentum value in laboratory system is defined with accuracy 0.05%. For that reason, the accurate knowledge of the^7H momentum vector together with information about only four of the five^7H →^3H + 4n decay products enables one to reconstruct the^7H excitation energy with high resolution.

The discussed approach will allow one to see the real profiles of the ground state (2.2 MeV) and the first excited state (5.5 MeV)^7H resonances with the FWHM resolution making 0.45 MeV and 0.75 MeV, respectively. These values are almost 2 times better than the resolutions achieved in [1, 2].

4.^3H nuclei and neutrons emitted at the^7H decay

The detection of^7H decay neutrons, their time-of flight and trajectory measurement, is the principal function of the Neutron Wall presented above. Operating together with the RIB diagnostic system of the ACCULINNA-2 separator the Neutron Wall placed at a distance 230 cm downstream the target will provide one-percent precision in the velocity measurement done for the neutrons. Their trajectory angles will be measured with accuracy ±20 mrad.

The 1.5-mm thick annular Si multi-strip detector 2b (see Figure 2), having the same sensitive area as the recoil detectors 2a, will give ΔE signal for the main part of the^3H nuclei emitted from the target offering the measurement of their specific energy loss (ΔE) and emission angle (θ_{lab} made with accuracy 150 keV and 7.5 mrad, respectively. Some of these^3H nuclei, emitted at the 7H decay at larger lab angles, will miss 2b. But information on their specific energy loss and emission angles will come from the ΔE − E array 2a.

The plastic detector array 4 is placed in front of NW (see this Figure 2) denoted in the whole setup and shown in front view in insert given in the Figure 2 right side. It is intended for measuring the rest energy E and the flight time (ToF) of^3H nuclei at the distance from the target making 225 cm. It is made of ten 1-meter long and 88-mm wide BC-404 scintillator plates having thickness equal 20 mm. Being disposed as two groups in five plates these plastics will be placed on the two sides of axis Y, marked in Figure 1. Each scintillator plate is viewed from two sides by the PMTs resulting in the definition of^3H stop time made with 0.2-ns accuracy. The^3H nuclei emitted from the target with energy ≤ 100 MeV are stopped in the 20 mm thick plastic.

The pair of 45 cm long, 20-mm thick plastics, each coupled with PMT on one side, will be placed in front of NW to cover its middle area left uncovered by the 1-meter plastics. This plastic pair will leave open free way for^8He nuclei to pass to the beam stopper. Thus, the ToF-ΔE-E identification will be provided for the^3H nuclei coming from the target. The ToF values measured with resolution 0.4 ns will give energy determination made for the 3H nuclei emitted at the five-body decay of^7H nucleus.

MC simulations were made for the^7H five-body decay in the framework of phase-space volume approximation. In particular, shown in Figure 5 are the MC simulated radial distributions obtained at the distance 225 cm from the target for the^7H decay tritons. Patterns in Fig. 6 allow one to see the energy values of the^3H nuclei shown against their emission angles. Presented in this figure are the simulation results where the detection of the^3H recoils, emitted at the reaction center-of-mass angles 0–20 degrees, is supplemented with the arrival of all^7H decay products,^3H and 4 neutrons, to the plastic array 4 and NW 5, respectively (see Figure 2).

Distributions simulated for the neutron arrival positions at the Neutron Wall are shown in Figure 7. Looking at the neutron energy spectra shown in Figure 8 one can see a quite large spread in the energy and ToF values.

5. Luminosity estimation

Basic for the estimations made on the setup efficiency is the chance that minimum four of the five^7H decay products arrive at the detectors 2a, 2b,

Figure 5. MC simulated radial distributions obtained at the distance 225 cm from the target for the^7H decay tritons. On the left and right sides shown are the distributions obtained for the decay of the ground (2.2 MeV) and first (5.5 MeV) 7H states, respectively, populated in the^2H(^8He,^3He)^7H reaction in the center-of-mass angular range 0 – 20 degrees.

Figure 6. Energy vs. angle distributions obtained by MC simulation for the^3H nuclei emitted at the 5-body decay of^7H nuclei produced in the^2H(^8He,^3He)^7H reaction in their 2.2 MeV ground state (left side) and in the 5.5 MeV first resonance state (right side).

Figure 7. MC simulated transverse profiles (left column) and radial distributions (right column) obtained at the Neutron Wall for the^7H decay neutrons emitted from the ground (2.2 MeV) and the first excited (5.5 MeV)^7H states.

Figure 8. The energy spectra of neutrons emitted from the deuterium target when 7H is produced in its ground (2.2 MeV) and the first excited (5.5 MeV) states in the^2H(^8He,^3He)^7H reaction. Given on top are the neutron ToF values calculated for the 230 cm distance to the Neutron Wall.

Table 2. Probabilities to have the^7H decay products arriving at the position in front of the Neutron Wall in coincidence with the^3He recoil detection. 7H states g. s., 2.2 MeV 1st state, 5.5 MeV 2nd state, 7.5 MeV θ_{cm} (deg) 0–20 20–30 0–20 20–30 0–20 20–30

3He + t4n 0.25 0.27 0.049 0.032 0.019 0.012 3He + t3n 0.32 0.36 0.29 0.22 0.18 0.14 3He + 4n 0.065 0.027 0.013 0.019 0.008 0.009

Table 3. Probabilities (ε_d) estimated for the^3He-triggered detection of minimum four particles emitted at the five-body^7H decay. 7H states g. s., 2.2 MeV 1st state, 5.5 MeV 2nd state, 7.5 MeV

θ_{cm} (deg)

0–20 20–30 0–20 20–30 0–20 20–30

3He + t4n t4n 0.0015 0.0017 0.0003 0.0002 0.0001 0.0001

t3n 0.022 0.024 0.0043 0.0028 0.0017 0.0011 3He + t3n 0.0070 0.0079 0.0064 0.0048 0.0040 0.0031

3He + 4n 4n 0.0004 0.0002 0.0001 0.0001 0.0000 0.0001 3He + 4n 3n 0.0057 0.0024 0.0011 0.0017 ε_d, all together

0.037 0.0362 0.0122 0.0096 0.0065 0.0052

4 and 5 (see Figure 2) when the^3He recoil emitted in the^2H(^8He,^3He)^7H reaction is detected. The results of such estimations made on the basis of MC simulation are given in Table 2.

Knowing these results and taking from Table 1 the values o f neutron flux decrease and depletion constant λ, the detection probabilities were estimated for the particle groups indicated in Table 2. These results are given in Table 3.

Detection probabilities presented in Table 3 (last row) offer the way to calculate the luminosity achievable in the experiment planned for the study of^7H energy spectrum populated in the transfer reaction^2H(^8He,^3He)^7H. Predictions made for the luminosity L appearing for the^7H formed in the three energy states, 2.2, 5.5, and 7.5 MeV are presented in Table 4.

Luminosity is calculated as the product L = ε_d × t × I × ΔΩ, with the detection probability ε_d given in Table 3, assumed target thickness t = 7.5 × 10^{20} cm^{−2}, the^8He beam intensity I = 2 × 10^5 s^{−1}, and with the estimated values of the solid angle covered by the triggering array 2a shown in Figure 2.

Table 4. Luminosity estimations.

7H states g. s., 2.2 MeV 1st state, 5.5 MeV 2nd state, 7.5 MeV θ_{cm} (deg) 0–20 20–30 0–20 20–30 0–20 20–30

ε_d 0.037 0.0362 0.0122 0.0096 0.0065 0.0052 ΔΩ, sr 0.35 0.08 0.33 0.27 0.31 0.46 L, cm^{−2}s^{−1} 8.5·10^{23} 2.2·10^{23} 3.0·10^{23} 2.0·10^{23} 1.5·10^{23} 1.8·10^{23}

Data reported in [3] allow for rough estimations made on the cross sections of the^2H(^8He,^3He)^7H reaction resulting in the population of the^7H ground state and its first excited state having energy approx2.2 MeV and approx5.5 MeV, respectively, above the^3H + 4ndecay threshold. Average differential cross section estimated for the^2H(^8He,^3He)^7H reaction populating the^7H ground state in angular range θ_{cm} = 0^∘˘20^∘ is 1×10^{−29} cm^2/sr. Such estimate made for the first excited state population is 3×10^{−29} cm^2/sr. Consequently, total statistics gained during the one-month time length of data taking done with the^8He beam intensity 2 × 10^5 s^{−1} on the target will contain 50 fivebody decay events recorded for each of these two^7H states.

6. Cross-talk and pile-up caused losses

The Neutron Wall composition made with the 75-mm thick plastics results that less than 5% of the total number of detected 3-fold and 4-fold neutron events are spoiled by the occurrence of such events when a single neutron is detected, due to re-scattering, in two adjacent scintillators (the so-called cross-talk). Time difference < 1 ns between the two neutron signals distinguishes these cross-talks from the overwhelming part of events obtained with two^7H decay neutrons detected in adjacent scintillators. Throwing out all events looking as cross-talks will not result in any noticeable loss in the luminosity of the experiment. The loss occurring from this ejection in the real collected data will be less than 2%.

Probability to have two neutrons hitting a single plastic makes about 8% when four neutrons emitted at the^7H decay arrive at the Neutron Wall. More than one-half of such pile-up events will be stored with the complete data set, characterizing the^3He recoil, and with four (3 neutrons and^3H) of the five^7H decay products. About 4% of pile-up events are possible when three neutrons, emitted at the^7H decay, arrive at the Neutron Wall. Therefore, conclusion is made that the neutron pile-up results in a 4% reduction of detection efficiency as compared with the data given in the last row of Table 3.

In summary, taking into account the cross-talk rejection and the losses caused due to pile-ups one should take luminosity estimates reduced by 6% as compared with the numbers presented in Table 4.

In addition, we expect a small background contribution from gamma rays. The main gamma sources could be materials 1, 2a, 2b and 3 (Figure 2). In this case, the time of flight to the Neutron Wall will be less than ∼ 9 ns, which is far from the typical time of flights corresponded to neutrons, see Figure 8.

7. Experimental test and conclusions

The given above estimations of cross-talk and detection efficiency, made for the Neutron Wall looking as a very compact assembly of BC-404 plastics, should be checked experimentally in a wide energy range of neutron energy 15–45 MeV. For this task the measurements of the^1H(^3H,^3He)n reaction, induced by the^3H beam obtained at ACCULINNA-2 with energy 25–60 MeV and intensity 10^6 s^{−1}, are foreseen. The recoil^3He nuclei are emitted from the target showing the neutron energy and flight direction. In the^1H(^3H,^3He)n reaction cross section making ∼ 1 mb/sr and at the thickness of hydrogen target ∼ 10^{21} cm^{−2} about 1 neutron per minute will be detected by a single BC-404 detector module included into an array of 7 plastics shown in Figure 1. In these measurements the triggering process (detection of low energy, ∼ 10−20 MeV,^3He recoil nuclei) using Si annular telescope will be optimized as well.

The proposed detector systems^1, together with the existing array of stilbene modules [8], will increase significantly the luminosity of the ACCULINNA- 2 setup which plays a key role in the experiments with radioactive beams [9]. These simulations, carried out for the^2H(^8He,^3He)^7H reaction, are the first approximation to the detail studies made on the multi-neutron decay of exotic nuclei like^7H,^7He^∗ and^6H, produced with the^8He beam bombarding deuterium target, and on the 4n excitation spectrum obtained in the 2H(^8He,^6Li)4n reaction.

Acknowledgments

This work was done in the frame of collaboration with National Center of Physics and Mathematics (project 8 “Physics of hydrogen isotopes”, topic 8.3 “Study of properties of neutron-rich nuclei away from the stability valley”). The activity was partly supported by the Russian Science Foundation Grant No. 22-12-00054. We acknowledge Prof. M. S. Golovkov for critical remarks and recommendations.

References

1. Bezbakh A.A., Chudoba V., Krupko S.A., Belogurov S.G., Biare D., Fomichev A.S., Gazeeva E.M., Gorshkov A.V., Grigorenko L.V., Kaminski G., Kiselev O.A., Kostyleva D.A., Kozlov M.Y., Mauyey B., Mukha

1Present status of the new detector array is as follows: 1) there are all components of NW, i. e. 100 BC-404 plastics and ETE-9822B photomultipliers; 2) the annular Si detectors with 32/125 mm inner/outer diameter segmented in 64 rings and 64 sectors will be ready in fall 2023; 3) the array of ten 1-meter long plastic scintillators, 88 mm wide and 20 mm thick, is under design.

I., Muzalevskii I.A., Nikolskii E.Y., Parfenova Y.L., Piatek W., Quynh A.M., Schetinin V.N., Serikov A., Sidorchuk S.I., Sharov P.G., Slepnev R.S., Stepantsov S.V., Swiercz A., Szymkiewicz P., Ter-Akopian G.M., Wolski R., Zalewski B., Zhukov M.V. Evidence for the First Excited State of^7H // Phys. Rev. Lett. — 2020. — 1. — V. 124. — P. 022502.

2. Muzalevskii I.A., Bezbakh A.A., Nikolskii E.Y., Chudoba V., Krupko S.A., Belogurov S.G., Biare D., Fomichev A.S., Gazeeva E.M., Gorshkov A.V., Grigorenko L.V., Kaminski G., Kiselev O., Kostyleva D.A., Kozlov M.Y., Mauyey B., Mukha I., Parfenova Y.L., Piatek W., Quynh A.M., Schetinin V.N., Serikov A., Sidorchuk S.I., Sharov P.G., Shulgina N.B., Slepnev R.S., Stepantsov S.V., Swiercz A., Szymkiewicz P., Ter-Akopian G.M., Wolski R., Zalewski B., Zhukov M.V. Resonant states in^7H: Experimental studies of the^2H(^8He,^3 He) reaction // Phys. Rev. C. — 2021. — 4. — V. 103. — P. 044313.

3. Nikolskii E.Y., Muzalevskii I.A., Bezbakh A.A., Chudoba V., Krupko S.A., Belogurov S.G., Biare D., Fomichev A.S., Gazeeva E.M., Gorshkov A.V., Grigorenko L.V., Kaminski G., Khirk M., Kiselev O., Kostyleva D.A., Kozlov M.Y., Mauyey B., Mukha I., Parfenova Y.L., Piatek W., Quynh A.M., Schetinin V.N., Serikov A., Sidorchuk S.I., Sharov P.G., Shulgina N.B., Slepnev R.S., Stepantsov S.V., Swiercz A., Szymkiewicz P., Ter- Akopian G.M., Wolski R., Zalewski B., Zhukov M.V.^6H states studied in the^2H(^8He,^4 He) reaction and evidence of an extremely correlated character of the^5H ground state // Phys. Rev. C. — 2022. — Jun. — V. 105. — P. 064605.

4. Kohley Z., Lunderberg E., DeYoung P., Roeder B., Baumann T., Christian G., Mosby S., Smith J., Snyder J., Spyrou A., Thoennessen M. Modeling interactions of intermediateenergy neutrons in a plastic scintillator array with Geant4 // Nuclear Instruments and Methods in Physics Research Section A: Accelerators, Spectrometers, Detectors and Associated Equipment. — 2012. — V. 682. — P. 59–65.

5. Tippawan U., Pomp S., Blomgren J., Dangtip S., Gustavsson C., Klug J., Nadel-Turonski P., Nilsson L., Österlund M., Olsson N., Jonsson O., Prokofiev A.V., Renberg P.U., Corcalciuc V., Watanabe Y., Koning A.J. Light-ion production in the interaction of 96 MeV neutrons with carbon // Phys. Rev. C. — 2009. — 6. — V. 79. — P. 064611.

6. Tilquin I., El Masri Y., Parlog M., Collon P., Hadri M., Keutgen T., Lehmann J., Leleux P., Lipnik P., Ninane A., Hanappe F., Bizard G., Durand D., Mosrin P., Péter J., Régimbart R., Tamain B. Detection efficiency of the neutron modular detector DEMON and related characteristics // Nuclear Instruments and Methods in Physics Research Section A: Accelerators, Spectrometers, Detectors and Associated Equipment. — 1995. — V. 365, no. 2. — P. 446–461.

7. Sharov P.G., Fomichev A.S., Bezbakh A.A., Chudoba V., Egorova I.A., Golovkov M.S., Golubkova T.A., Gorshkov A.V., Grigorenko L.V., Kaminski G., Knyazev A.G., Krupko S.A., Mentel M., Nikolskii E.Y., Parfenova Y.L., Pluchinski P., Rymzhanova S.A., Sidorchuk S.I., Slepnev R.S., Stepantsov S.V., Ter-Akopian G.M., Wolski R. Search for 2p decay of the first excited state of^{17}Ne // Phys. Rev. C. — 2017. — Aug. — V. 96. — P. 025807.

8. Bezbakh A.A., Belogurov S.G., Wolski R., Gazeeva E.M., Golovkov M.S., Gorshkov A.V., Kaminski G., Kozlov M.Y., Krupko S.A., Muzalevky I.A., Nikolskii E.Y., Ovcharenko E.V., Slepnev R.S., Ter-Akopian G.M., Fomichev A.S., Chudoba V., Sharov P.G., Schetinin V.N. A Neutron Spectrometer for Experiments with Radioactive Beams on the ACCULINNA-2 Fragment Separator // Instruments and Experimental Techniques. — 2018. — 9. — V. 61, no. 5. — P. 631–638.

9. Fomichev, A. S.., Grigorenko, L. V.., Krupko, S. A.., Stepantsov, S. V.., Ter-Akopian, G. M.. The ACCULINNA-2 project: The physics case and technical challenges // Eur. Phys. J. A. — 2018. — V. 54, no. 6. — P. 97.