arXiv:1807.11005v1 [astro-ph.IM] 29 Jul 2018

Kyoto’s Event-Driven X-ray Astronomy SOI pixel sensor for the FORCE mission

Takeshi G. Tsuru^a, Hideki Hayashi^a, Katsuhiro Tachibana^a, Sodai Harada^a, Hiroyuki Uchida^a, Takaaki Tanaka^a, Yasuo Arai^b, Ikuo Kurachi^b, Koji Mori^c, Ayaki Takeda^c, Yusuke Nishioka^c, Nobuaki Takebayashi^c, Shoma Yokoyama^c, Kohei Fukuda^c, Takayoshi Kohmura^d, Kouichi Hagino^d, Kenji Ohno^d, Kohsuke Negishi^d, Keigo Yarita^d, Shoji Kawahito^e, Keiichiro Kagawa^e, Keita Yasutomi^e, Sumeet Shrestha^e, Shunta Nakanishi^e, Hiroki Kamehama^f, and Hideaki Matsumura^g

aDepartment of Physics, Graduate School of Science, Kyoto University, Kitashirakawa, Sakyo-ku, Kyoto, 606-8502, Japan bInstitute of Particle and Nuclear Studies, High Energy Accelerator Research Org., KEK, 1-1 Oho, Tsukuba 305-0801, Japan cDepartment of Applied Physics and Electronic Engineering, Faculty of Engineering, University of Miyazaki, 1-1 Gakuen Kibanadai-Nishi, Miyazaki, 889-2192, Japan dDepartment of Physics, Faculty of Science and Technology, Tokyo University of Science, 2641 Yamazaki, Noda, Chiba 278-8510, Japan eResearch Institute of Electronics, Shizuoka University, Johoku 3-5-1, Naka-ku, Hamamatsu 432-8011, Japan fInformation and Communication Systems Engineering, Okinawa National College of Technology, Henoko 905, Nago, Okinawa, Japan gKavli IPMU, The University of Tokyo, 5-1-5 Kashiwanoha, Kashiwa, Chiba, 277-8583, Japan

ABSTRACT We have been developing monolithic active pixel sensors, X-ray Astronomy SOI pixel sensors, XRPIXs, based on a Silicon-On-Insulator (SOI) CMOS technology as soft X-ray sensors for a future Japanese mission, FORCE (Focusing On Relativistic universe and Cosmic Evolution). The mission is characterized by broadband (1-80 keV) X-ray imaging spectroscopy with high angular resolution (< 15 arcsec), with which we can achieve about ten times higher sensitivity in comparison to the previous missions above 10 keV. Immediate readout of only those pixels hit by an X-ray is available by an event trigger output function implemented in each pixel with the time resolution higher than 10 μsec (Event-Driven readout mode). It allows us to do fast timing observation and also reduces non-X-ray background dominating at a high X-ray energy band above 5–10 keV by adopting an anti-coincidence technique. In this paper, we introduce our latest results from the developments of the XRPIXs. (1) We successfully developed a 3-side buttable back-side illumination device with an imaging area size of 21.9 mm×13.8 mm and an pixel size of 36 μm × 36 μm. The X-ray throughput with the device reaches higher than 0.57 kHz in the Event-Driven readout mode. (2) We developed a device using the double SOI structure and found that the structure improves the spectral performance in the Event-Driven readout mode by suppressing the capacitive coupling interference between the sensor and circuit layers. (3) We also developed a new device equipped with the Pinned Depleted Diode structure and confirmed that the structure reduces the dark current generated at the interface region between the sensor and the SiO_2 insulator layers. The device shows an energy resolution of 216 eV in FWHM at 6.4 keV in the Event-Driven readout mode.

Keywords: SOI Pixel Sensors, X-rays, Imaging, Spectroscopy, FORCE mission

Further author information: (Send correspondence to T.G.T.) T.G.T.: E-mail: tsuru@cr.scphys.kyoto-u.ac.jp, Telephone: 81 75 753 3868

1. INTRODUCTION: THE FORCE MISSION We are now proposing a future Japan-lead X-ray mission, FORCE, to be realized in 2026.^{1, 2} The primary scientific objective of the FORCE mission is to hunt for “missing black holes” in various mass-scales and to trace their cosmic evolution. The black holes include super-massive black holes in highly-obscured AGNs, intermediate massive black holes which might be the seeds for super-massive black holes, and isolated stellar mass black holes. We identify and quantify families of black holes whose populations and evolutions are unknown. Through this, we explore the evolution paths of the Universe.

Broadband observation is essential to uncover buried AGNs, to measure the masses of the black holes and to distinguish isolated stellar mass black holes from neutron stars and white dwarfs. The FORCE mission offers broadband (1-80 keV) X-ray imaging spectroscopy with high angular resolution < 15 arcsec in HPD, which offers about ten times higher sensitivity than NuSTAR at 1-80 keV.^3 For this purpose, FORCE has 3 identical pairs of super-mirrors and detectors. X-ray mirror is Light-weight Si mirror provided by NASA/GSFC.^4 We adopt a hybrid type of detector to cover the broadband, whose overall design is based on the HXI onboard the Hitomi satellite.^5 The detector imager consists of Si and CdTe layers. In this report, we introduce the latest results from the development of X-ray SOI pixel sensors as the Si layer.

2. XRPIX: EVENT DRIVEN X-RAY SOI PIXEL SENSOR Reduction of non-X-ray background (NXB) due to interactions of cosmic-rays is essential to detect faint X-ray objects, especially in the energy band above 10 keV. We adopt the anti-coincidence technique with scintillators surrounding the X-ray imager to reduce NXB. Since the counting rate of the scintillators is estimated to be around ∼ 10 kHz in a low earth orbit,^{6, 7} the time resolution of X-ray CCD is too poor to use this technique. On the other hand, the double-sided Si strip detector (DSSD) can not detect X-rays below ∼ 2 keV due to its relatively high readout noise, although the time resolution is high enough to adopt the anti-coincidence technique. This is the motivation for developing the X-ray astronomy SOI pixel sensors.

The SOI pixel sensor is monolithic, using a bonded wafer of a high resistivity depleted Si layer for X-ray detection, a SiO_2 insulator layer (buried oxide: BOX) and a low resistivity Si layer for CMOS circuits. “XRPIXs” refer to the X-ray astronomy SOI pixel sensors. In order to realize the time resolution significantly higher than ∼ 100 μsec (1/10 kHz), each pixel of an XRPIX is equipped with its own trigger logic circuit and analogue readout CMOS circuit, which is the most important feature of the XRPIX. Table 1 shows the target specification of the XRPIX for the FORCE mission.

Figure 1 shows the in-pixel CMOS circuit implemented in each pixel and the peripheral readout circuit. The in-pixel CMOS circuitry consists of an analog readout circuit and an trigger output circuit. The signal charge from the PN diode is converted to signal voltage at the charge sensitive amplifier (CSA). The correlated double sampling (CDS) circuit, consisting of the CDA capacitance and CDS reset transistor, cancels the reset noise

Table 1. Target Specification of the XRPIX

Item Specification Imaging Imaging Area: 15 × 45 mm^2, Pixel Size: 36 μm (0.74^{′′} at F= 10 m) Bandpass 1–40 keV (Requirement), 0.3–40 keV (Goal) Back side dead layer thickness 1 μm (Requirement), 0.1 μm (Goal) Depletion layer thickness 200 μm (Requirement), 500 μm (Goal) Energy resolution at 6 keV (FWHM) 300 eV (Requirement), 140 eV (Goal) Equivalent noise charge (rms) 10 e (Requirement), 3 e (Goal) Time resolution 10 μsec Throughput 2 kHz (counting rate of the Crab nebula) Non-X-ray background at 20 keV 1/100 of X-ray CCD (5 × 10^{−5} cps/keV/10 × 10mm^2) generated at the CSA circuit. The signal voltage is read out through the source follower (SF) implemented in the pixel, and the peripheral readout circuit consisting of the column programable gain amplifier and output buffer. We use an inverter-chopper type of comparator in the trigger output circuit. The threshold level is set at VTH through the VTH reset transistor. The trigger output signal can be masked at pixel level. We read out 8 × 8 pixels for each event to see the charge sharing and to collect all the signal charge produced by an X-ray event.

Figure 1. In-pixel CMOS circuit and peripheral readout circuit of XRPIX5b having an n-type sensor layer.

The XRPIX has two readout modes. One is “Frame readout mode”, in which all the pixels are read out serially like a CCD without using the trigger function. The other is “Event-Driven readout mode”, in which the hit pixel and its surrounding 8 × 8 pixels, with the hit pixel in the center, are read out using the trigger function. Figure 2 shows how to read out an X-ray event in the Event-Driven readout mode. (1) Assuming an X-ray is detected at the pixel labeled with “X-ray !” in Figure 2, the pixel trigger circuit writes its address into the row and column hit resisters. (2) A trigger flag (TRIG O) is immediately sent to the FPGA. (3) The FPGA reads the address of the hit pixel. (4) The FPGA writes the address of the pixel to be read into the row and column readout registers, (5) then reads out the analog signal of the pixel with the ADC.

Figure 2. The readout sequence in the Event-Driven readout mode.

3. RESULTS FROM RECENT DEVELOPMENTS 3.1 Imaging in the Event Driven Readout Mode XRPIX5b is a 3-side buttable back-illumination device and the largest XRPIX sensor we have processed. The device has an imaging area size of 21.9 mm×13.8 mm, an pixel size of 36 μm × 36 μm and a format of 608 × 384. The sensor layer is n-type Si with a resistivity of ∼ 5 Ωcm.

Figure 3 shows a demonstration of a Cd-109 X-ray image obtained with XRPIX5b operated in the Event- Driven readout mode.^8 We placed the openwork of the clock tower of Kyoto University as the mask over the XRPIX5b, and applied a back bias voltage of 10 V at room temperature. The thickness of the depletion layer is estimated to be ∼ 130 μm. Also, the device successfully detected Am-241 X-rays at a count rate of ∼ 570 Hz without using the mask. The back bias voltage was 10 V and the operating temperature was −60^∘C. Since the count rate was limited by the RI source used in the experiment, the maximum throughput would be higher than ∼ 570 Hz.

Figure 3. Demonstration of X-ray imaging performance of XRPIX5b in the Event-Driven readout mode. The right panel shows a Cd-109 X-ray image obtained with XRPIX5b in the Event-Driven readout at room temperature. The left panel is a photo of the openwork used in the demonstration as the mask (the clock tower of Kyoto university).

3.2 Double-SOI structure Figure 4 shows the evolution of the spectral performance in the Frame readout mode from 2009 to 2017. The spectral performance in the Frame readout mode became significantly better by reducing the parasitic capacitance at the sense node, by introducing the charge sensitive amplifier instead of the source follower and by developing new device structures.

Figure 5 shows the spectra obtained with an identical device of XRPIX3b in the Frame and Event-Driven modes, respectively.^9 The spectral performance in the Event-Driven readout mode was significantly worse than that in the Frame readout mode.^{10} In the Event-Driven readout mode, the energy resolution is poor and there is an offset in the output channel. The difference between the two modes is whether the in-pixel digital circuit is used or not. Therefore, it is conceivable that the interference with the analog signal due to the operation of the digital circuit degrades the spectral performance in the Event-Driven readout mode

We found two causes degrading the spectral performance in the Event-Driven readout mode. First is that the analog and digital circuits have a common power supply line, resulting in the common impedance coupling. We already modified the power lines in the latest device so that each circuit has its own designated power line. The other is crosstalk between the digital circuit and the buried well (n-type buried well is applied for the p-type Si sensor layer), which is electrically connected to the sense-node. Figure 6 (a) shows the cross-sectional views of the single SOI structure, which we have been using so far. There is the capacitive coupling between the BNW (buried n-well) and the trigger signal line in the circuit layer.^{10}

Figure 4. Evolution of the spectral performance of X-ray astronomy SOIPIX in the Frame readout.^{9, 11–13}

Figure 5. The spectra of the Am-241 X-rays obtained with an XRPIX3b (left) in the Frame readout mode and (right) in the Event-Driven readout mode.^9

In order to solve the interference problem, we adopt a Double-SOI wafer (DSOI) in which we introduce an additional Si layer (middle Si layer) between the circuit and sensor layers.^{14} The BOX layer is interleaved with the middle Si layer,^{15–17} as shown in Figure 6 (b). Note that the sensor layer of this Double-SOI device is p-type whereas that of the XRPIX5b is n-type. The middle Si layer is expected to act as an electrostatic shield and to reduce the capacitive coupling between the BNW and the digital circuit.

Figure 7 shows the spectra of Co-57 X-ray we obtained with an XRPIX6D having the Double-SOI structure in the Frame and Event-Driven readout modes. The performance in the Event-Driven mode is significantly improved by adopting the Double-SOI structure and is now close to that in the Frame readout mode. The

Figure 6. Cross-sectional views of (a) the single SOI and (b) Double-SOI structures.^{15–17} A p-type Si sensor layer is assumed in this figure.

energy resolutions are 312 eV and 346 eV at 6 keV in the Frame and Event-Driven readout modes, respectively. No significant offset in the output channel is observed in the Event-Driven readout mode. The results show that the crosstalk between the circuit and sensor layers is suppressed as expected. We found that the sense-node gain is increased by about a factor of two in comparison to the single SOI device having the same design of the in-pixel CSA. This should be due to the reduction in the sense-node parasitic capacitance by making the area of the BNW smaller, and also to the increase in the closed-loop gain by reducing the feedback parasitic capacitance between the CSA and the BNW.^{13, 18, 19}

Figure 7. Co-57 spectra obtained with the Double-SOI device (XRPIX6D) (a) in the Frame readout mode and (b) in the Event-Driven readout mode.^{13}

3.3 Pinned Depleted Diode structure In the single SOI structure, the charge generated in the interface region between the sensor and BOX layers is collected, which results in a significantly large dark current degrading the spectral performance. The device also suffers from the possibility of signal charge loss by the traps at the interface, which results in the degradation of the charge collection efficiency.^{20} This situation is unchanged even with the Double-SOI structure.

In order to solve these problems, Kamehama et al. (2018) recently developed a Pinned Depleted Diode (PDD) structure.^{12} The PDD structure has a BPW region beneath the BOX layer, and a BNW region below that in the single SOI wafer as shown in Figure 8. The signal charge generated by an X-ray collects through the stepped buried n-well (BNW1, BNW2 and BNW3) into the readout node (n^+) without touching the interface between the sensor (p–) and the BOX layers. Thus, the signal charge loss by the traps at the interface does not occur.

The sufficiently highly doped buried p-well (BPW1) as a neutral region reduces the dark current generation from the interface. It works just like a pinned photodiode in CCD or CMOS image sensors. The buried p-well acts as a shielding layer between the sensor and circuit layers, and suppresses the interference between the two layers in the same way as the middle Si layer in the Double-SOI structure.

Figure 9 shows the dark current per pixel as a function of working temperature.^{12} The dark current generation of the device with the PDD structure is two orders of magnitude lower than that of the device with the single SOI at 25^∘C. A dark current value of 10 e msec^{−1} pixel^{−1} can be achieved by cooling the device to below −10^∘C. We note that the temperature dependency tends to saturate at a temperature lower than −5^∘C. Kamehama et al. (2018) suggest trap-assisted band-to-band tunneling as the possible reason for the saturation.^{12}

Figure 8. Cross-sectional view of SOIPIX-PDD. The figure is adopted from Kamehama et al. (2018).^{12}

Figure 9. Dark current per pixel as a function of working temperature in the single SOI device (Conventional SOIPIX) and in the PDD device (Proposed SOIPIX-PDD). The figure is adopted from Kamehama et al. (2018).^{12}

The device with the PDD structure exhibits a better spectral performance than that with the single SOI structure as well as that with the Double-SOI structure. Figure 4 (f) shows the X-ray spectrum of Fe-55 obtained with single pixel in the device (SOIPIX-PDD) having the PDD structure developed by Kamehama et al. (2018).^{12} The energy resolution of 200 eV in FWHM at 5.9 keV and the ENC of 11 e in rms are the best values ever obtained with the X-ray SOI pixel sensors, although this device does not support the Event-Driven readout mode. Figure 10 shows the X-ray spectra of Co-57 and Am-241 obtained in the Event-Driven readout mode with a new device, XRPIX6E, which is the first PDD device supporting the Event-Driven readout mode.^{21}

While the spectra were collected from single arbitrarily chosen pixel, it was already confirmed that the device exhibits the same spectral performance when using other single pixel. The energy resolution is 216 eV at 6.4 keV, which is close to that obtained in the Frame readout mode.

Figure 10. X-ray spectra of Co-57 (left) and Am-241 (right) obtained in the Event-Driven readout mode with XRPIX6E.^{21}

4. SUMMARY AND FUTURE PROSPECTS After about 10 years of development, we have successfully realized a device with a sensor size of 21.9 mm×13.8 mm and a format of 608 × 384, and achieved a time resolution higher than 10 μsec and a throughput higher than 570 Hz in the Event-Driven readout mode. We also successfully reduced the interference between the sensor and circuit layers by improving the device structure. The latest device shows an energy resolution of 216 eV in FWHM at 6.4 keV in the Event-Driven readout mode.

In order to realize a high spectral performance and a large format at the same time, we are now developing a new device with an imaging area size of 21.9 mm × 13.8 mm having the Double-SOI or PDD structure. While the energy resolution at or above 6 keV was significantly improved, the X-ray performance at a low X-ray energy band below 2 keV is still insufficient for the FORCE mission.^{22} Therefore, we will study the backside process using the latest devices, XRPIX6D and XRPIX6E, and make necessary improvements.

ACKNOWLEDGMENTS We acknowledge the valuable advice and great work by the personnel of LAPIS Semiconductor Co., Ltd. This study was supported by the Japan Society for the Promotion of Science (JSPS) KAKENHI Grant-in-Aid for Scientific Research on Innovative Areas 25109002 (Y.A.), 25109003 (S.K.), 25109004 (T.G.T., T.T., K.M. and A.T.), 20365505 (T.K), 23740199 (T.K), 18740110 (T.K.), Grant-in-Aid for Young Scientists (B) 15K17648 (A.T.), Grant-in-Aid for Challenging Exploratory Research 26610047 (T.G.T.) and Grant-in-Aid for JSPS Fellows 15J01842 (H.M.). This study was also supported by the VLSI Design and Education Center (VDEC), the University of Tokyo in collaboration with Cadence Design Systems, Inc., and Mentor Graphics, Inc.

REFERENCES [1] Mori, K., Tsuru, T. G., Nakazawa, K., Ueda, Y., Okajima, T., Murakami, H., Awaki, H., Matsumoto, H., Fukazawa, Y., Tsunemi, H., Takahashi, T., and Zhang, W. W., “A broadband x-ray imaging spectroscopy with high-angular resolution: the FORCE mission,” in [Space Telescopes and Instrumentation 2016: Ultraviolet to Gamma Ray], den Herder, J.-W. A., Takahashi, T., and Bautz, M., eds., 99051O, International Society for Optics and Photonics (Aug. 2016).

[2] Nakazawa, K., Mori, K., Tsuru, T. G., Ueda, Y., Awaki, H., Fukazawa, Y., Ishida, M., Matsumoto, H., Murakami, H., Okajima, T., Takahashi, T., Tsunemi, H., and Zhang, W. W., “The FORCE mission: science aim and instrument parameter for broadband x-ray imaging spectroscopy with good angular resolution,” in [Space Telescopes and Instrumentation 2018: Ultraviolet to Gamma Ray], den Herder, J.-W. A., Nakazawa, K., and Nikzad, S., eds., 106992D, International Society for Optics and Photonics (July 2018).

[3] Harrison, F. A., Craig, W. W., Christensen, F. E., Hailey, C. J., Zhang, W. W., Boggs, S. E., Stern, D., Cook, W. R., Forster, K., Giommi, P., Grefenstette, B. W., Kim, Y., Kitaguchi, T., Koglin, J. E., Madsen, K. K., Mao, P. H., Miyasaka, H., Mori, K., Perri, M., Pivovaroff, M. J., Puccetti, S., Rana, V. R., Westergaard, N. J., Willis, J., Zoglauer, A., An, H., Bachetti, M., Barrière, N. M., Bellm, E. C., Bhalerao, V., Brejnholt, N. F., Fuerst, F., Liebe, C. C., Markwardt, C. B., Nynka, M., Vogel, J. K., Walton, D. J., Wik, D. R., Alexander, D. M., Cominsky, L. R., Hornschemeier, A. E., Hornstrup, A., Kaspi, V. M., Madejski, G. M., Matt, G., Molendi, S., Smith, D. M., Tomsick, J. A., Ajello, M., Ballantyne, D. R., Baloković, M., Barret, D., Bauer, F. E., Blandford, R. D., Brandt, W. N., Brenneman, L. W., Chiang, J., Chakrabarty, D., Chenevez, J., Comastri, A., Dufour, F., Elvis, M., Fabian, A. C., Farrah, D., Fryer, C. L., Gotthelf, E. V., Grindlay, J. E., Helfand, D. J., Krivonos, R., Meier, D. L., Miller, J. M., Natalucci, L., Ogle, P., Ofek, E. O., Ptak, A., Reynolds, S. P., Rigby, J. R., Tagliaferri, G., Thorsett, S. E., Treister, E., and Urry, C. M., “THE NUCLEAR SPECTROSCOPIC TELESCOPE ARRAY( NuSTAR) HIGH-ENERGY X-RAY MISSION,” The Astrophysical Journal 770, 103 (June 2013).

[4] Zhang, W. W., Allgood, K. D., Biskach, M. P., Chan, K.-W., Hlinka, M., Kearney, J. D., Mazzarella, J. R., McClelland, R. S., Numata, A., Riveros, R. E., Saha, T. T., and Solly, P. M., “Astronomical x-ray optics using mono-crystalline silicon: high resolution, light weight, and low cost,” in [Space Telescopes and Instrumentation 2018: Ultraviolet to Gamma Ray], den Herder, J.-W. A., Nakazawa, K., and Nikzad, S., eds., 106990O, International Society for Optics and Photonics (July 2018).

[5] Nakazawa, K., Sato, G., Kokubun, M., Enoto, T., Fukazawa, Y., Hagino, K., Hayashi, K., Kataoka, J., Katsuta, J., Kobayashi, S. B., Laurent, P., Lebrun, F., Limousin, O., Maier, D., Makishima, K., Mizuno, T., Mori, K., Nakamori, T., Nakano, T., Noda, H., Odaka, H., Ohno, M., Ohta, M., Saito, S., Sato, R., Tajima, H., Takahashi, H., Takahashi, T., Takeda, S., Tanaka, T., Terada, Y., Uchiyama, H., Uchiyama, Y., Watanabe, S., Yamaoka, K., Yatsu, Y., and Yuasa, T., “Hard x-ray imager onboard Hitomi (ASTRO-H),” Journal of Astronomical Telescopes, Instruments, and Systems 4, 021410 (Apr. 2018).

[6] Takahashi, T., Abe, K., Endo, M., Endo, Y., Ezoe, Y., Fukazawa, Y., Hamaya, M., Hirakuri, S., Hong, S., Horii, M., Inoue, H., Isobe, N., Itoh, T., Iyomoto, N., Kamae, T., Kasama, D., Kataoka, J., Kato, H., Kawaharada, M., Kawano, N., Kawashima, K., Kawasoe, S., Kishishita, T., Kitaguchi, T., Kobayashi, Y., Kokubun, M., Kotoku, J., Kouda, M., Kubota, A., Kuroda, Y., Madejski, G., Makishima, K., Masukawa, K., Matsumoto, Y., Mitani, T., Miyawaki, R., Mizuno, T., Mori, K., Mori, M., Murashima, M., Murakami, T., Nakazawa, K., Niko, H., Nomachi, M., Okada, Y., Ohno, M., Oonuki, K., Ota, N., Ozawa, H., Sato, G., Shinoda, S., Sugiho, M., Suzuki, M., Taguchi, K., Takahashi, H., Takahashi, I., Takeda, S., Tamura, K.-i., Tamura, T., Tanaka, T., Tanihata, C., Tashiro, M., Terada, Y., Tominaga, S., Uchiyama, Y., Watanabe, S., Yamaoka, K., Yanagida, T., and Yonetoku, D., “Hard X-Ray Detector (HXD) on Board Suzaku,” Publications of the Astronomical Society of Japan 59, S35–S51 (Jan. 2007).

[7] Kokubun, M., Makishima, K., Takahashi, T., Murakami, T., Tashiro, M., Fukazawa, Y., Kamae, T., Madejski, G. M., Nakazawa, K., Yamaoka, K., Terada, Y., Yonetoku, D., Watanabe, S., Tamagawa, T., Mizuno, T., Kubota, A., Isobe, N., Takahashi, I., Sato, G., Takahashi, H., Hong, S., Kawaharada, M., Kawano, N., Mitani, T., Murashima, M., Suzuki, M., Abe, K., Miyawaki, R., Ohno, M., Tanaka, T., Yanagida, T., Itoh, T., Ohnuki, K., Tamura, K.-i., Endo, Y., Hirakuri, S., Hiruta, T., Kitaguchi, T., Kishishita, T., Sugita, S., Takahashi, T., Takeda, S., Enoto, T., Hirasawa, A., Katsuta, J., Matsumura, S., Onda, K., Sato, M., Ushio, M., Ishikawa, S.-n., Murase, K., Odaka, H., Suzuki, M., Yaji, Y., Yamada, S., Yamasaki, T., Yuasa, T., and the HXD team, “In-Orbit Performance of the Hard X-Ray Detector on Board Suzaku,” Publications of the Astronomical Society of Japan 59, S53–S76 (Jan. 2007).

[8] Hayashi, H., Tsuru, T. G., Tanaka, T., Uchida, H., Matsumura, H., Tachibana, K., Harada, S., Takeda, A., Mori, K., Nishioka, Y., Takebayashi, N., Yokoyama, S., Fukuda, K., Arai, Y., Kurachi, I., Kawahito, S., Kagawa, K., Yasutomi, K., Shrestha, S., Nakanishi, S., Kamehama, H., Kohmura, T., Hagino, K., Negishi, K., Oono, K., and Yarita, K., “Evaluation of Kyoto’s Event-Driven X-ray Astronomical SOI Pixel Sensor with a Large Imaging Area,” Nuclear Instruments and Methods in Physics Research Section A: Accelerators, Spectrometers, Detectors and Associated Equipment, submitted (2018).

[9] Takeda, A., Tsuru, T. G., Tanaka, T., Uchida, H., Matsumura, H., Arai, Y., Mori, K., Nishioka, Y., Takenaka, R., Kohmura, T., Nakashima, S., Kawahito, S., Kagawa, K., Yasutomi, K., Kamehama, H., and Shrestha, S., “Improvement of spectroscopic performance using a charge-sensitive amplifier circuit for an X-ray astronomical SOI pixel detector,” Journal of Instrumentation 10, C06005–C06005 (June 2015).

[10] Takeda, A., Tsuru, T., Tanaka, T., Matsumura, H., Arai, Y., Mori, K., Nishioka, Y., Takenaka, R., Kohmura, T., Nakashima, S., Kawahito, S., Kagawa, K., Yasutomi, K., Kamehama, H., and Shrestha, S., “Development and Evaluation of Event-Driven SOI Pixel Detector for X-ray Astronomy,” Proceedings of Technology and Instrumentation in Particle Physics 2014 (TIPP2014). 2-6 June 2014. Amsterdam, the Netherlands. Online at http://pos.sissa.it/cgi-bin/reader/conf.cgi?confid=213, id.138 TIPP2014 (2014).

[11] Ryu, S. G., Tsuru, T. G., Nakashima, S., Takeda, A., Arai, Y., Miyoshi, T., Ichimiya, R., Ikemoto, Y., Matsumoto, H., Imamura, T., Ohmoto, T., and Iwata, A., “First Performance Evaluation of an X-Ray SOI Pixel Sensor for Imaging Spectroscopy and Intra-Pixel Trigger,” IEEE Transactions on Nuclear Science 58, 2528–2536 (Oct. 2011).

[12] Kamehama, H., Kawahito, S., Shrestha, S., Nakanishi, S., Yasutomi, K., Takeda, A., Tsuru, T. G., and Arai, Y., “A Low-Noise X-ray Astronomical Silicon-On-Insulator Pixel Detector Using a Pinned Depleted Diode Structure,” Sensors 18(1), 27 (2018).

[13] Takeda, A., Mori, K., Nishioka, Y., Fukuda, K., Hida, T., Yukumoto, M., Tsuru, T. G., Tanaka, T., Uchida, H., Matsumura, H., Hayashi, H., Tachibana, K., Harada, S., Arai, Y., Kurachi, I., Kohmura, T., Hagino, K., Yarita, K., Negishi, K., Oono, K., Kawahito, S., Kagawa, K., Yasutomi, K., Shrestha, S., and Kamehama, H., “Improvement of spectroscopic performance with Double-SOI pixel detectors for an X-ray astronomy,” in Prep. (2018).

[14] Ohmura, S., Tsuru, T. G., Tanaka, T., Uchida, H., Takeda, A., Matsumura, H., Ito, M., Arai, Y., Kurachi, I., Miyoshi, T., Nakashima, S., Mori, K., Nishioka, Y., Takebayashi, N., Noda, K., Kohmura, T., Tamasawa, K., Ozawa, Y., Sato, T., Konno, T., Kawahito, S., Kagawa, K., Yasutomi, K., Kamehama, H., Shrestha, S., Hara, K., and Honda, S., “Reduction of cross-talks between circuit and sensor layer in the Kyoto’s X-ray astronomy SOI pixel sensors with Double-SOI wafer,” Nuclear Instruments and Methods in Physics Research Section A: Accelerators, Spectrometers, Detectors and Associated Equipment 831, 61–64 (Sept. 2016).

[15] Miyoshi, T., Arai, Y., Chiba, T., Fujita, Y., Hara, K., Honda, S., Igarashi, Y., Ikegami, Y., Ikemoto, Y., Kohriki, T., Ohno, M., Ono, Y., Shinoda, N., Takeda, A., Tauchi, K., Tsuboyama, T., Tadokoro, H., Unno, Y., and Yanagihara, M., “Monolithic pixel detectors with 0.2 μm FD-SOI pixel process technology,” Nuclear Instruments and Methods in Physics Research Section A: Accelerators, Spectrometers, Detectors and Associated Equipment 732, 530–534 (Dec. 2013).

[16] Honda, S., Hara, K., Tsuchida, K., Asano, M., Tobita, N., Maeda, T., Arai, Y., Miyoshi, T., Go Tsuru, T., Ohno, M., Miura, N., Kasai, H., and Okihara, M., “Total Ionization Damage Compensations in Double Silicon-on-Insulator Pixel Sensors,” Proceedings of Technology and Instrumentation in Particle Physics 2014 (TIPP2014). 2-6 June 2014. Amsterdam, the Netherlands. Online at http://pos.sissa.it/cgibin/reader/conf.cgi?confid=213, id.138 TIPP2014 (2014).

[17] Hara, K., Arai, Y., Honda, S., Kurachi, I., Tsuboyama, T., Tobita, N., Miyoshi, T., Asano, M., and Mitsui, S., “Initial Characteristics and Radiation Damage Compensation of Double Silicon-on-Insulator Pixel Device,” PoS Vertex2014, 033 (May 2015).

[18] Miyoshi, T., Arai, Y., Fujita, Y., Hara, K., Ikegami, Y., Kurachi, I., Tauchi, K., Tsuboyama, T., Yamada, M., Ono, S., Nishimura, R., and Hamasaki, R., “Front-end electronics of double SOI X-ray imaging sensors,” Journal of Instrumentation 12, C02004–C02004 (Feb. 2017).

[19] Miyoshi, T., Arai, Y., Fujita, Y., Hamasaki, R., Hara, K., Ikegami, Y., Kurachi, I., Nishimura, R., Ono, S., Tauchi, K., Tsuboyama, T., and Yamada, M., “Performance study of double SOI image sensors,” Journal of Instrumentation 13, C02005–C02005 (Feb. 2018).

[20] Matsumura, H., Tsuru, T. G., Tanaka, T., Takeda, A., Arai, Y., Mori, K., Nishioka, Y., Takenaka, R., Kohmura, T., Nakashima, S., Hatsui, T., Kohmura, Y., Takei, D., and Kameshima, T., “Improving chargecollection efficiency of SOI pixel sensors for X-ray astronomy,” Nuclear Instruments and Methods in Physics

Research Section A: Accelerators, Spectrometers, Detectors and Associated Equipment 794, 255–259 (Sept. 2015).

[21] Harada, S., Tsuru, T. G., Tanaka, T., Uchida, H., Matsumura, H., Tachibana, K., Hayashi, H., Takeda, A., Mori, K., Nishioka, Y., Takebayashi, N., Yokoyama, S., Fukuda, K., Arai, Y., Kurachi, I., Kawahito, S., Kagawa, K., Yasutomi, K., Shrestha, S., Nakanishi, S., Kamehama, H., Kohmura, T., Hagino, K., Negishi, K., Oono, K., and Yarita, K., “Investigation of Event Driven X-ray Astronomical SOI Pixel Sensor, XRPIX with Pinned Depleted Diode Structure,” Nuclear Instruments and Methods in Physics Research Section A: Accelerators, Spectrometers, Detectors and Associated Equipment, submitted (2018).

[22] Itou, M., Tsuru, T. G., Tanaka, T., Takeda, A., Matsumura, H., Ohmura, S., Uchida, H., Nakashima, S., Arai, Y., Kurachi, I., Mori, K., Takenaka, R., Nishioka, Y., Kohmura, T., Tamasawa, K., and Tindall, C., “The first back-side illuminated types of Kyoto’s X-ray astronomy SOIPIX,” Nuclear Instruments and Methods in Physics Research Section A: Accelerators, Spectrometers, Detectors and Associated Equipment 831, 55–60 (Sept. 2016).