arXiv:1204.3415v5 [physics.ins-det] 13 Aug 2012

SPHERES, Jülich’s High-Flux Neutron Backscattering Spectrometer at FRM II

Joachim Wuttke,^∗ Alfred Budwig, Matthias Drochner, Hans Kämmerling, Franz-Joseph Kayser, Harald Kleines, Vladimir Ossovyi, Luis Carlos Pardo,^† Michael Prager,^‡ Dieter Richter, Gerald J. Schneider, Harald Schneider, and Simon Staringer Forschungszentrum Jülich GmbH, 52425 Jülich, Germany

SPHERES (SPectrometer with High Energy RESolution) is a third-generation neutron backscattering spectrometer, located at the 20 MW German neutron source FRM II and operated by the Jülich Centre for Neutron Science. It offers an energy resolution (fwhm) better than 0.65 μeV, a dynamic range of ±31 μeV, and a signal-to-noise ratio of up to 1750:1.

This preprint is outdated. Please use only the published version: Rev. Sci. Instrum. 83, 075109 (2012); http://dx.doi.org/10.1063/1.4732806; also available from

http://apps.jcns.fz-juelich.de/spheres

I. INTRODUCTION

Neutron backscattering is a versatile technique for measuring the dynamics of spins, atoms, and molecules on a GHz scale. Typical applications include hyperfine interactions in magnetic materials, molecular rotations, diffusion, and relaxation in complex systems.^{1,2}

Historically, three instrument generations can be distinguished. Proposed by Maier-Leibnitz in 1966,^3 neutron backscattering was first realized in a test setup at the 4 MW reactor FRM (Garching, Germany). Count rates were of the order of one per hour, and the signal-to-noise ratio was 2:1 at best so that it took several weeks to measure a single spectrum.^4 Following this demonstration of principle, first-generation backscattering spectrometers were build at Jülich (“π-Spektrometer”^5) and Grenoble (IN10^{6–8}, IN13^9); they were reviewed in Refs. 1,10–12.

In the second-generation instrument IN16 of Institut Laue-Langevin (ILL) at Grenoble, consequent use has been made of focussing optics; a deflector chopper allows to operate a spherical monochromator in exact backscattering geometry.^{13–15} Another instrument of this type is planned to be built at the Australian research reactor OPAL.^{16}

In third-generation spectrometers, the focussing optics is made even more efficient by a phase-space transform (PST) chopper. This device, invented by Schelten and Alefeld in 1984,^{17} enhances the neutron spectrum available at the monochromator, and allows for a particularly compact instrument design. The first realisation has been the High-Flux Backscattering Spectrometer (HFBS) at the National Institute of Standards and Technology (NIST, Gaithersburg, Md.).^{18,19}

In this article, we present SPHERES, the SPectrometer for High Energy RESolution built and operated by JCNS (Jülich Centre for Neutron Science) at FRM II (Forschungsneutronenquelle Heinz Maier-Leib-nitz of Technische Universität München at Garching, Germany). A third PST based backscattering instrument is currently under construction at the ILL (project IN16B) where it will replace both IN10 and IN16.^{20,21}

The development of SPHERES has been documented in several publications.^{22–27} First productive experiments were performed in June 2007, and routine operation started in April 2008. Since then, two thirds of the beam time is offered to external users. Experiment proposals are invited twice a year through the JCNS user office. By now (March 2012), nearly one hundred experiments have been performed, and more than twenty publications have appeared. In the following, we describe the design of SPHERES, its hardware and software components, and its performance characteristics. The supplemental material contains a list of suppliers and some technical details.^{29}

II. INSTRUMENT DESIGN

A. Instrument Layout

In a backscattering spectrometer, strong energy filtering of incident and scattered neutrons is achieved by Bragg reflection from perfect monochromator and analyzer crystals under angles close to 180^∘. This peculiar geometry leads to some complications in the spectrometer design; it entails the need for a primary beam deflector and a duty-cycle chopper. A particularly compact design is achieved if these two functions are realized by one and the same component: by a chopper that bears deflector crystals on its circumference. As an important side-effect, the Bragg deflection by moving crystals results in a PST that enhances the useable flux at the monochromator.

The resulting instrument layout is shown in Fig. 1. It is similar to the layout of HFBS,^{19} with three major differences: (1) HFBS is operated under vacuum whereas the SPHERES housing is gas tight but not vacuum proof; (2) at HFBS the monochromator is located outside the instrument housing; (3) in the secondary spectrometer detectors and analyzers have changed sides with respect to the primary beam, resulting in a larger scattering angle range at SPHERES.

FIG. 1: Basic layout of SPHERES.

Bragg reflection for energy selection Si(111) Si(311) final neutron wavelength λ_f (Å) 6.271 3.275 final neutron energy E_f (meV) 2.080 7.626 final neutron velocity v_f (m/s) 630.9 1208 Bragg reflection for deflection PG(002) PG(004) deflection angle 2α0 (^∘) 41.65 25.14 Schelten-Alefeld velocity v0 (m/s) 300.1 290.5

TABLE I: Parameters that depend on the choice of Bragg reflections for energy selection and for deflection. The current configuration of SPHERES uses silicon (111) in monochromator and analyzers and pyrolitic graphite (002) as deflector.

B. Crystal Choices

The basic design decision for a backscattering spectrometer is the choice of the monochromator and analyzer crystals. Initially, SPHERES was foreseen to be convertible between a Si(111) and a Si(311) configuration. Si(111) is the standard at most backscattering spectrometers. The shorter wavelength of Si(311) gives access to a larger range in the scattering wave number q, with promising application niches in the study of diffusion on very short length scales.^{30,31} However, the lower resolution (which goes with λ^{−2}) and background problems from Bragg reflections (for instance from sample holders or from cryostat walls) have so far prevented wide-spread useage. In commissioning SPHERES, the original mechanics had to be modified in many details in order to block neutronic background channels and to provide biological shielding. As a result, the instrument has lost its convertibility; the Si(311) option has been abandoned for the foreseeable future.

Another important choice is pyrolitic graphite (PG) for the deflector crystals. Once deflector, monochromator, and analzyer crystals are chosen, the neutronic parameters listed in Table I are fixed. The angle 2α_0 between incoming and deflected beam is obtained from the Bragg equation 2 cos α_0 = λ/(c/l) where c = 6.710 Å is the lattice constant of PG, and l the order of the Bragg re-

FIG. 2: Neutron path versus time-of-flight: (a) As realized in SPHERES, with chopper opening frequency f1 = v_f /4R. (b) Originally planned for SPHERES (and realized in HFBS), with f2 = 3f1. The upward arrows describe neutrons that are directly scattered from the sample into the detector.

flection [00l].

C. Time-of-Flight Logic

Another decisive hardware choice is the radius R = 2000 mm of the analyzer spheres. This parameter fixes the time of flight τ := 2R/v_f = 6.34 ms for a round trip from the sample to the analyzer and back. During this round trip, the detectors see n pulses of directly scattered neutrons, interlaced with n − 1 pulses of energy-analyzed neutrons. If both types of pulses have the same duration τ_n, then τ_n = τ/(2n − 1), with n = 1, 2, . . . Accordingly, the chopper must open with a frequency f_n = 1/(2τ_n).

In the primary spectrometer, neutrons are deflected by the closed chopper. After a round trip to the monochromator and back, they must hit the open chopper phase. Therefore, the distance chopper – monochromator L_{CM} must obye 2L_{CM}/v_f = (2m − 1)τ_n. Since the monochromator should have about the same curvature as the analyzers, the only practicable choice is L_{CM} = R and m = n. All this is illustrated by the time-distance diagrams in Fig. 2.

HFBS uses n = 2, and so will IN16B. The present chopper of SPHERES also was originally designed for n = 2, but because of mechanical instabilities it is only operated with n = 1. This reduces the efficiency of the PST, but it has a positive effect upon the signal-to-noise ratio, to be explained below (Sect. VI C).

III. INSTRUMENT COMPONENTS

A. Neutron Guide and Velocity Selector

SPHERES is located in the neutron guide hall West of FRM II. Like all instruments in this hall, it receives neutrons through beam tube SR1 from the cold source. SPHERES uses an end position at the secondary neutron guide NL 6a. This guide has a cross-

FIG. 3: Deflector crystal assembly in the PST chopper. In both drawings, the neutrons come from the left. Left drawing: cross section. Right drawing: view from outside towards the chopper axis (the closing Al element and the screws are not shown). The deflector stacks consist of three pyrolitic graphite mosaic crystals, tilted by aluminum wedges, and followed by a^{10}B4C absorber plate.

section of 60 × 120 mm^2; its curved segments have a radius of 1000 m. The guide is shared with two upstream instruments, MIRA^{28} and DNS, that use thin crystal monochromators to extract neutrons at shorter wavelengths than those needed at SPHERES.

The instrument shutter of SPHERES and its mechanical neutron velocity selector are located in two separate biological shieldings. The velocity selector (EADS Astrium) reduces the incoming neutron spectrum to about 6.27±0.34 Å. It is followed by a convergent neutron guide (“anti-trumpet”, by S–DH) that reduces the beam section by a factor of 11.5 to 25 × 25 mm^2. Neutrons leave this guide system with a divergence of about ±47 mrad horizontal, ±64 mrad vertical. For details on all this, see the Supplemental Material.^{29}

B. Phase-Space Transform Chopper

The phase-space transform (PST) chopper is the key component of a third-generation backscattering spectrometer. As said above, it combines the functionalities of a beam deflector and of a duty-cycle chopper. Half of its circumference bears deflector crystals that redirect incoming neutrons towards the monochromator. The other half circumference is open. When the chopper is open, neutrons coming out of the neutron guide are transmitted towards a beam stop that is integrated in the chopper housing, and neutrons coming back from the monochromator are transmitted towards the sample.

To enable spectral scans, a sufficiently large wavelength band must be forwarded towards the monochromator. This requires deflector crystals with large mosaicity. Pyrolitic graphite (PG) crystals, specified for a mosaicity of 2.5^∘ fwhm (Advanced Ceramics) were selected

FIG. 4: Chopper operation mode, (a) originally intended, (b) currently realized. The white square shows the illuminated area. The arrows indicate the rotation during τ = 6.34 ms; in this time, deflected neutrons travel from the chopper to the monochromator and back.

according to rocking curves measured one-by-one on the triple-axis neutron spectrometer UNIDAS at Jülich. To further enhance the horizontal mosaicity, the crystals are assembled in stacks of three, with tilts of 0^∘, +2.5^∘, and −2.5^∘ imposed by aluminum wedges (Fig. 3).

Intensity and width of the forwarded neutron band are further enhanced by the horizontal motion of the deflector crystals. This phase-space transform ‘from white to wide’ has first been suggested in a conference proceedings of limited circulation;^{17} an easily accessible summary has been provided in Ref. 19. The Bragg reflection by moving mosaic crystals compresses the energy distribution and thereby enhances the flux in the acceptance range of the backscattering monochromator. In accordance with Liouville’s theorem, the compression of the energy distribution is accompanied by a widening of the angular distribution. This is not a severe drawback as long as the angular resolution of the entire instrument is limited by that of the secondary spectrometer.

Assuming an almost collimated incoming beam, an optimum transform is predicted for a deflector speed of

v_0 = v_f sin α_0 cos 2α_0 (1) with the neutron velocity v_f and the PG deflection angle α_0 as given in Tab. I. For Si(111) and PG(002), one obtains v_0 = 300.1 m/s. In the meantime, several simulations^{19,32,33} and a measurement^{19} indicate that under realistic conditions the optimum deflector speed is only about 250 m/s.

The current chopper rotor has a radius of 650 mm; the crystals are located at a mean radius of R = 600 mm. The rotor has k = 3 crystal bearing wings of 60^∘ each. Originally, the chopper was intended to operate with a pulse fragmentation index of n = 2 (Fig. 4a): Neutrons that have made the round trip chopper – monochromator – chopper would hit the second but next opening following the wing by which they had been deflected. The rotation frequency f_{rot} = (2n − 1)f_n/k = f_2 = 78.9 Hz would result in a deflector speed of 2πRf_{rot} = 297m/s.

Tensile forces for such R and f_{rot} are considerable. To keep them within admissible limits, the high-strength aluminum alloy alumold (ALZNMg2Mn) was chosen, and the geometry of the disk was optimized with finiteelement calculations, resulting in a construction with three spokes and two recesses for each of the three wings.^{34}

Thermal instabilities of the bearings and mechanical resonances prevent us from running the current rotor with the intended 78.9 Hz. As a fallback solution, we are operating the chopper with n = 1 and f_{rot} = f_1/3 = 26.3 Hz (Fig. 4b). A new rotor is currently under construction; it will have an asymmetric rotor (k = 1) to achieve a higher crystal speed while maintaining n = 1. As an engineering compromise, a deflector velocity of 225 m/s has been chosen, somewhat below the optimum, which, according to the simulations, is quite flat.

C. Monochromator and Analyzers

In the standard configuration, the Si(111) reflection is used in monochromator and analyzers. Single-crystalline hexagonal wafers with a side length of 58.5 mm and a thickness of 500 μm were glued onto spherical supports with a curvature radius of R = 2000 mm. The bending stress induces a lattice-constant gradient that significantly improves the backscattered intensity for a moderate cost in resolution.

The latter can be estimated quantitatively as follows. The relative change of the lattice constant at a distance d from the neutral surface is κd/R with a material constant κ = 0.776 for Si(111).^{35} Assuming a rectangular distribution of lattice constants, and neglecting Darwin tails and other dynamical scattering effects, one would expect a rectangular distribution of backscattered energies with a full width of ΔE = 0.81 μeV. Convoluting this distribution with itself, to account for monochromator and analyzer, one obtains a triangular distribution with the same full width ΔE. This is in reasonable agreement with the observed overall resolution of SPHERES, which is in first approximation a Gaussian with a full width at half maximum of about 0.62 μeV (Sect. VI D).

An alternate monochromator/analyzer set uses polished wafers with a cut-in grid supposed to make them stress free. The resolution is about 0.4 μeV, but the flux is so much lower that this configuration is of little use, except possibly for the study of narrow tunneling or hyperfine splitting.

Additional monochromators offer the possibility of measuring with a fixed energy offset. A CaF_2(111) monochromator, providing an offset of 24 μeV with respect to the Si(111) analyzers, is already available.

The monochromator support, made of carbon-fiber reinforced plastics, has a size of 500×250 mm^2 and a spherical curvature of about R + L_{CS}/2 = 2180 mm, where L_{CS} = 360 mm is the distance between the chopper and the nominal sample location, so that the monochromator acts as a hollow mirror that maps the chopper’s PG crystals onto the sample.

The small-angle analyzers consist of eight massive aluminum rings mounted on a common support. Each ring has a width of 95 mm; between two rings there is a gap of 10 mm. The innermost ring, with an inner radius of 165 mm, proved useless because it receives fringe intensity from the direct beam; therefore it was covered with Cd. The other seven rings cover scattering angles 2ϑ from 9.0 ± 1.4^∘ to 27 ± 1.4^∘. Typically, two rings focus into one detector.

The large-angle analyzers consist of seven shells, made of cast aluminum and finished by turning They have a height of 2000 mm, and a maximum width (in the scattering plane) of about 530 mm. One shell covers a solid angle of about 60^∘ × 15^∘ ≃ 0.27 rad^2. The seven shells are located at scattering angles from 30^∘ to 135^∘, with only small gaps between them. There is no one-to-one correspondence between individual shells and detectors; the seven shells must rather be seen as one contiguous surface.

D. Monochromator Doppler Drive

A novel, resonance-free monochromator drive has been developed jointly by Forschungszentrum Jülich and an industrial company (Aerolas). The development costs were shared with the ILL, which received an identical drive for IN16 and for future use with IN16B. The monochromator shell is mounted on an arbor that is supported by two air bearing collars. Driven by a linear motor, it performs a sinusoidal motion. The displacement amplitude Z can be chosen between ±25 and ±75 mm. The maximum frequency is about F = 10 Hz, resulting in a maximum velocity amplitude of V = 2πF Z = 4.7 m/s.

To give an idea of the engineering difficulties, let us note that the peak acceleration is 2πF V = 295 m/s^2. The moving mass is nearly 10 kg, and the maximum mechanical power is nearly 7 kW. The linear motors generate a heat flux of up to 2.5 kW that is removed by water cooling. The peak current of each of the two motors is about 70 A at 600 V; much of this electric power is recovered during deceleration.

We define the monochromator velocity v to take positive values if the monochromator is moving towards the sample. In the rest frame of the monochromator, backscattered neutrons have the velocity v_f = 630.9 m/s. In the laboratory frame, they have the energy

E_i(v) = m_n 2 (v_f + v)^2 . (2)

The analyzers select scattered neutrons with energy E_f = E_i(0). In consequence, the sample energy gain in the scattering process is

ℏω = E_i − E_f = ( 2 v v_f + v^2 v_f^2 ) E_f . (3)

The quadratic term causes a small asymmetry in the dynamic range. With V = 4.7 m/s, accessible energy transfers ℏω extend from −30.9 to 31.2 μeV.

E. Backscattering Detectors

The detector bank for large scattering angles consists of eleven counter tubes 24NH15, bought from Eurisys, filled with 4 bar of^3He. They have a diameter of 1 inch and a height of 210 mm.

The tubes are mounted vertically. They are alternately arranged on two circles with radii of 110 and 130 mm at intervals 10^∘, so that complete coverage is achieved for an angular range of 110^∘ in the scattering plane. In front of the tubes (starting at a circle with radius 80 mm around the ideal sample centre) there are radial, vertical Cd collimator plates. The vertical opening of the detector block has been reduced to about 80 mm in order to block neutrons that do not come from the analyzers.

For small scattering angles, front-window counters 10NH3/5X P (also from Eurisys) are used, filled with 3.5 bar of^3He. Their diameter is 2 inch, the active length is 1 inch. They are mounted around the exit nose of the chopper housing.

F. Diffraction Detectors

SPHERES is equipped with position-sensitive detector tubes (Reuter-Stokes) for auxiliary diffraction measurements to be performed simultaneously with the spectral scans. The tubes are located in the horizontal scattering plane in front of the analyzers; inevitably, they shield a few percent of the effective analyzer surface. At present, six tubes, with a diameter of 1 inch and 535 mm of active length, provide coverage of scattering angles from about 30 to 120^∘.

The resolution of this diffractometer is limited by the horizontal spread of incident neutron directions, which in the worst case is ±(half monochromator width)/R = ±250/2000 = ±12.5%. This is acceptable because we do not aim for flull-fledged structure determination but only for monitoring structural changes.^{36}

G. Instrument Housing with Argon Filling

The instrument housing has a surface of about 36 m^2, and a volume of nearly 60 m^3. Its walls and ceiling are made of steel plates of 5 mm thickness, anchored to a steel frame, and proven to be gas tight. Towards the inside, a 100 mm polyethylen layer moderates fast neutrons. The innermost cladding of walls and ceiling consists of Cd plates of 1 mm thickness.

The housing is reasonably gas tight and equipped with pneumatics and controls so that it can be filled with argon. This reduces losses by air scattering in the secondary spectrometer, where neutrons have to travel about 4.3 m through atmosphere. In the primary spectrometer, another 0.3 m are in atmosphere; the remaining fligh path is in a vacuum chamber that prolongates the chopper casing towards the monochromator, as indicated in Fig. 1. Displacing 90% of the air by argon leads to an intensity gain of more than 35%, in good accord with expectations from tabulated neutron cross sections. For technical details of the argon filling, see the Supplemental Material.^{29}

H. Sample Environments

The standard sample environment, used for most experiments, is a cryostat SHI-950T (Janis) It is a closedcycle refrigerator system with cold head and compressor manufactured by Sumitomo Heavy Industries. In the low-temperature range from 4 K to 320 K, the sample is cooled via exchange gas. In the high-temperature range, from 290 K to 650 K (700 K for limited amount of time), the system acts as a cryofurnace, with the sample in vacuum. The sample tube has a diameter of 60 mm; samples are allowed to have diameters of up to 55 mm.

The standard sample stick is equipped with Si diodes DT670A for temperatures from 4 K to 500 K. A hightemperature stick with Pt 1000 thermometers covers temperatures from 10 K to 700 K. Gas-loading equipment is under preparation.

Sample environments are inserted and operated from the platform on top of the instrument housing. If the tail diameter exceeds 160 mm, the secondary spectrometer must be temporarily realigned to a sample position slightly out of the focus of the monochromator. To change the sample environment, it is unavoidable to break the argon atmosphere. Therefore, experiments with non-standard sample environments are preferentially scheduled at the beginning or at the end of a reactor cycle. So far, experiments have been performed with a dilution cryostat that gives access to temperatures down to 30 mK. Other environments will be adapted and tried as requested by users.

IV. INSTRUMENT CONTROL AND DATA ACQUISITION

A. High-Level User Interface

During regular experiments, physical access to the instrument is only needed for changing samples. All other operations can be performed remotely. A graphical user interface (GUI) allows

• to open and close the instrument shutter,

• to start and stop the monochromator drive, and to set its velocity amplitude,

• to set the sample temperature, and to start temperature ramps,

• to start and stop the data acquisition, to save files, and to reset histograms,

• to submit, edit, start and stop experiment scripts.

It displays

• warnings and error messages,

• the status of data acquisition, sample environment, and peripheral systems,

• current count rates for all detectors, separately for direct, elastic, and inelastic scattering,

• the latest acquired spectrum,

• the latest acquired diffractogram,

• the time evolution of various parameters in the last hours or days,

• the status of experiment-script execution and the script commands ahead.

As the instrument is operated from different workplaces (cabin in the neutron guide hall, office, home), there is no simple way to maintain a traditional log book. Therefore, we have replaced paper by digital storage, using a simple wiki system^{37} as our instrument log. Besides, there are automatically generated event and parameter logs.

B. Instrument-Control Daemons

As a leading principle in the design of our instrumentcontrol software, we requested that no neutrons should be lost because of peripheral software or network problems. Therefore, the instrument’s subsystems are controlled by a number of independent background processes (daemons) that continue to work even if the connection to upper-level processes is interrupted.

The following daemons are the most important ones:

• spectral data acquisition (Sect. IV D),

• diffraction data acquisition (Supplemental Material)^{29}

• slow control daemon (Sect. IV E),

• temperature-controller controller, supervising the sample-environment temperature controller,

• Doppler driver, supervising the control PC supplied by the manufacturer,

• event logger, receiving log lines from other daemons and writing them to an event log, thereby documenting user interventions, experiment-script execution steps, system warnings and error messages.

FIG. 5: The data acquisition VME crate.^{38} The signal processor on the SIS 3100 sequencer board is a ADSP21061. An optical fiber connects the crate to a SIS 1100 PCI card on the dedicated data-acquisition PC.

• parameter logger, querying every 30 s some fifty status parameters from other daemons and writing them to a parameter log,

• documentation server and raw-data postprocessor (Sect. V A),

• script engine, executing user-written experiment scripts,

• GUI server, through which data and commands are channeled to and from the graphical user interface.

The spectral acquisition is time critical; therefore it is written in the programming language C, and it is running on dedicated computer. For all other processes, computational speed is less important than clarity of code; therefore they are written in an expressive scripting language, Ruby.

All computers are connected by ethernet (instrument LAN). Interprocess communication (IPC) is based on TCP (Transmission Control Protocol) messages. Each daemon is listening on a TCP port for commands given by other daemons. For instance, the spectral acquisition daemon receives commands (start, stop, save, reset) from the GUI server, and status queries from the parameter logger.

C. Spectral Acquisition Electronics

The data acquisition is done in a VME crate (Fig. 5) using components and software from Struck Innovative Systeme (SIS).^{38} The input consists of TTL pulses from the neutron-counter preamplifiers and from the optical encoders sensing the translation of the monochromator

Doppler drive (200 ticks per mm) and the rotation of the PST chopper (20480 ticks per revolution). The Doppler and chopper ticks are counted up and down in homemade counter boards, then they are transmitted to a 32 bit latch. The neutron counts are transmitted via a trivial home-made interface board to a 32×32 scaler.

The sampling period is τ = 20 μs. Once every τ, a digital signal processor reads the data from the latch and from the scaler, writes them to DRAM, and resets the scaler. The data form a fixed-length block of 20 32-bit words. In the following, each block shall be identified by an index i, corresponding to a time t_i = const + iτ. One word encodes the Doppler position z_i and the chopper phase χ_i (16 bits each); the remaining 19 words n_{ji} accomodate the neutron counts from up to 19 detectors j. The DRAM is divided into two buffers with 1’000’000 words each. It takes 50’000 data blocks or 1 s to fill one buffer. While one buffer is being filled, the other buffer is read out via an optical link by the upstream PC.

D. Spectral Acquisition Daemon

The data-acquisition PC (Opteron 2×2.4 GHz, 64bit), is dedicated to running the spectral acquisition daemon. It accumulates and stores histograms containing neutron counts per energy channel and per detector. These raw spectra are only a few correction steps (Sect. V B) away from the physical scattering law S(q, ω).

The daemon consists of an endless loop that waits for one of the two DRAM buffer on the VME crate to be filled. As soon as this is the case, its contents is read via the optical link and pushed on a stack. When the stack covers at least one Doppler oscillation period, it is processed. A sine function with four free parameters (offset, amplitude, frequency, phase) is fitted to the z_i, using starting values retained from the previous fit. Then, a loop over the data blocks i is performed, to be described in the next three paragraphs.

Depending on the chopper phase χ_i, the data block is assigned a category c(χ_i), which can be indirect, direct, or ambiguous, depending on whether or not the neutrons have travelled the detour sample–analyzer–sample or not. In Sect. VI C we will show how this categorization is set up. Ambiguous data blocks are not processed further.

To assign our scattering events to an energy channel ω, we need to know the time

t̂_i = t_i − (v_f + v(t̂_i))(L^0_i − z(t̂_i)) − v_f L^c_f (4)

at which the neutron has been backscattered by the monochromator. Here, t_i is the detection time, and L_{i,f} is the path travelled before and after scattering by the sample. The path L^c_f depends on the category c: it is only about 10 cm in the direct case, but 4 m more in the indirect case. The path offset z(t̂_i) and the velocity modulation v(t̂_i) are given by position and speed of the monochromator at time t̂_i. This makes Eq. (4) nonlinear; it is most easily solved by iteration. For a given t̂_i, the analytic derivative of the fit function z(t) is used to obtain v(t̂_i). Eq. (3) then yields ω(v_i). A quick estimate shows that the delay t̂_i − t_i is not negligible: Neutrons in the indirect category travel more than 6 m from the monochromator to the detector, which takes about 10 ms, corresponding to up to 1/10 of a Doppler period.

On starting the data acquisition for a given Doppler velocity amplitude, an equidistant ω mesh has been chosen, with channels [ω_k − Δω/2, ω_k + Δω/2]. This makes it easy to determine the pertinent channel k(i) = ⌊(ω(v_i) − (ω_0 − Δω/2))/Δω⌋. Finally, the neutron histograms of the pertinent chopper category c(χ_i) are incremented: H^c_{jk} ← H^c_{jk} +n_{ji} for all detectors j. The time spent in energy channel k(i) is incremented accordingly: N^c_k ← N^c_k + 1. Here ends the loop over i.

All these computations must be completed before the next DRAM buffer is overwritten. After some optimizations, this requirement could be fulfilled with sufficient margin: In years of operation, not a single data block has been lost because of computer overload. As a side product, a self-contained C library for Levenberg-Marquardt least-squares minimization has been released as an opensource project,^{39} serving practitioners from all areas of science and engineering.

Our histogramming relies on the assumption that the motion of the monochromator accurately follows a simple analytic function. Analyses of recorded z_i(t_i) traces have shown that the motion accomplished by the Aerolas Doppler drive is indeed almost perfectly sinusoidal, deviations being negligible on the scale set by the energy resolution of the spectrometer.

The accumulated histograms are written to files when a preset time interval has elapsed or when a save command is received through the daemon’s IPC interface. Files are saved to the local disk, which is remote readable through NFS (Network File System). The rawspectra, as other data produced by SPHERES, are formatted as self-documenting YAML files. YAML is a data-serialization specification with intrinsic support for basic data structures like arrays and hashes and with an emphasis on human readability; parsers are available for all major programming languages.^{40}

E. Safety Instrumentation and Slow Control

SPHERES is equipped with a fail-safe programmable logic controller (Fehlersichere speicherprogrammierbare Steuerung F-SPS, Siemens Simatic S7 300F) for safety instrumentation and slow control.^{41} The F-SPS is connected via internal bus or Profibus-DP to various sensors and actors. A touchpanel next to the instrument door provides a status display; it provides the only regular user interface for safety-critical subsystems like the door lock and the argon pneumatics.

A simple daemon running on a compact-PCI computer couples the F-SPS to the instrument LAN. It has read access to all sensors and write access to safety-uncritical actors, including the shutter opener, which can be activated if the instrument housing is closed and locked and a manual switch is in “remote control” position.

V. DATA PROCESSING

A. Raw-Data Postprocessor

The raw files produced by the spectral data-acquisition daemon contain the energy mesh ω_k, the histograms H^c_{jk} and N^c_k , some information about the instrument and data-acquisition setup, and the date and time when the measurement was started and stopped. Had all other daemons crashed, this would be sufficient to allow a valid data analysis. In regular operation, however, users expect data files to contain additional information like experiment title, sample name, and sample temperature. Since the spectral acquisition daemon is deliberately kept ignorant about these things (not essential for its functioning, and potential sources of bugs and failures), the wanted information must be added ex post.

This is done by the raw-data postprocessor. Running on another computer than the spectral acquisition daemon, it reads a raw-spectrum file via NFS, and looks up the start and stop time entries. Then, it opens the parameter log, reads the lines recorded during the spectral measurement, and computes mean value and standard deviation of the sample temperature and of some other parameters. Textual information like experiment title and sample name, usually entered by the user via the GUI, is retrieved from another dedicated file system. All this is written to a user-friendly aggregated raw-data file.

B. Raw-Data Reduction with SLAW

While some users want to work on the raw data, most ask for standard software to deliver normalized scattering laws S(q, ω). To provide this service, a new raw-data reduction program Slaw^{42} has been developed. Compared to legacy software like SQW of the ILL, Slaw offers better support for four-dimensional data sets S(q, ω; T ) or S(q, ω; t) obtained in time-resolved inelastic scans. Also, Slaw does not bind users to a specific data format; it supports a variety of output formats, and new output routines can be easily added ad hoc.

Slaw is controlled by an input script that contains configuration commands and a map that assigns raw scans to output files. Slaw processes the raw data through the following steps of which all but the first and the last are optional:

• read raw data files, adding or subtracting subscans as needed;

• perform binning in energy, scattering angle, or time slots, possibly eliminating some of these dimensions;

• subtract empty-can measurement, optionally weighted with a self-absorption factor;^{43}

• normalize neutron counts H^c_{jk} to the measuring time N^c_k τ spent in a given energy channel;

• correct for incident flux variations (Sect. V C);

• divide counts by elastic intensity of a reference measurement (usually vanadium or a low-temperature measurement with the current sample);

• save the scattering law.

Slaw has alternative input routines to process raw data from other backscattering spectrometers. Its structure is generic enough to allow future extension to timeof-flight spectrometers.

C. Instead of a Monitor

Most inelastic neutron spectrometers have a hightransmission neutron counter, located a little upstream of the sample, to monitor the incoming flux. In SPHERES, the windows of such a monitor would cause undesirable background. We therefore prefer an indirect normalization.

We must account for two kinds of flux variation: temporal variations of the reactor power, of the filling of the cold source, and of the monochromator orientations of the upstream instruments, and spectral variations of the primary spectrometer transmission. The temporal variations of the incident beam can be measured with good accuracy without inserting any device into the beam: it is sufficient to monitor the γ radiation emitted by the aluminum windows at the end of the convergent guide and at the chopper entry. Such a γ monitor, still missing, will be installed soon.

The spectral variation of the flux at the sample is determined by spectrum of the cold source, by the neutron guide system, by the velocity selector, and most importantly, by the PST chopper and the monochromator. To make things more complicated, the angle of the Bragg deflection in the PST chopper is energy dependent, and the monochromator is moving in space so that the geometric-optical focussing onto the sample is imperfect, with a non-trivial interplay of energy, direction, and lateral spread. To account for all this, the spectral distribution of the incident flux is best measured either by placing a neutron counter exactly at the sample position, or by counting neutrons that are scattered from the sample directly into the regular detectors. The latter method is regularly used as the “pseudomonitor” of SPHERES.

VI. PERFORMANCE

A. Flux and Count Rates

The incident flux at the end of the convergent neutron guide has been measured at two occasions with gold foil activation. The results were (1.85 ± 0.05) · 10^9 cm^{−2}s^{−1} in November 2005, and (1.19 ± 0.05) · 10^9 cm^{−2}s^{−1} in February 2012. A decrease of the order of 10% was to be expected from the two breaks in the upstream neutron guide (Sect. III A). The remaining loss is not understood; it must be feared that the neutron guides have degraded substantially. To detect further degradation and to determine necessary action, it will be important to remeasure the flux more regularly in the coming years.

The flux at the sample position is more difficult to measure; for details see the Supplemental Material.^{29} The best current estimate is ϕ_{sam} = 1.8 · 10^6 s^{−1} within a a cross-section of 25 × 40 mm^2. Here and in the following, fluxes are reported for the open chopper state. The average flux seen by a gold foil is only half of it.

For typical samples made for transmissions between 80% and 90%, elastic count rates ϕ_{el} are of the order 1000 s^{−1}. Here as in the instrument’s GUI display, such numbers are per detector, and under the assumption that all time is spent counting into this single channel. The pseudomonitor rate ϕ_{dir} of neutrons that are directly scattered from the sample into the detector, is about twice as large. Reasons why ϕ_{el}/ϕ_{dir} is smaller than 1 include: solid angle mismatch (the detector block has a larger solid angle than the analyzers, to ensure that all backscattered neutrons have a chance to reach a detector), absorption and scattering losses in the sample (which must traversed a second time by backscattered neutrons), losses along the 4 m from the detector to the analyzers and back, imperfect energy selection in the monochromator, finite backscattering probability in the analyzers.

B. Dark Counts

Dark count rates in each of the large-angle detectors are of the order of 0.02 s^{−1} when the reactor is off, and 0.1 s^{−1} when the reactor is at 20 MW and the intrument shutter closed. This is negligible compared to the background produced in our primary spectrometer.

When the instrument shutter is open and the chopper is running, but the exit window of the chopper (just before the sample position) is closed by a Cd mask, then the dark count rates in the large-angle detectors are about 0.53 s^{−1}. Only a small part of this background is due to slow neutrons that penetrate from the primary into the secondary spectrometer, circumventing the chopper exit window. Most of the background is independent of the chopper phase, which indicates an origin between the shutter and the chopper. This has been confirmed by experimental tests. Probably, a n–γ conversion in the supermirror coating of the focussing neutron guide is followed by a conversion to fast neutrons in the guide’s lead shielding.

C. Chopper Modulation of Count Rates

Along with the duty-cycle chopper, a backscattering spectrometer needs a time-of-flight logic to discriminate neutrons that are backscattered by the analyzers from neutrons that are scattered directly from the sample into the detectors. In SPHERES, this time-of-flight discrimination is performed by the spectral acquisition daemon (Sect. IV D), based on the chopper phase reading that is saved every 20 μs along with the Doppler position reading and the neutron counts.

FIG. 6: Neutron counts versus chopper phase. Data from the same low-temperature resolution measurement as in Fig. 7. In the direct phase, neutrons are scattered without energy analysis from the sample into the detectors. In the indirect phase, registered neutrons come from the analyzer. If the scattering is purely elastic, as in this example, then the inelastic signal is just undesired background.

The parametrization of the discrimination must be determined empirically. A chopper-phase histogram is saved in each raw data file. This histogram is based on a rectangular grid with 120 chopper-phase channels, but only two energy channels, to distinguish roughly between elastic and inelastic scattering. Fig. 6 shows such data from a resolution measurement. There is a range of nearly 60^∘ with very high count rates, independently of the Doppler velocity, due to direct scattering. There is another range, labelled indirect, where almost all neutrons have been backscattered by an analyzer, as can be inferred from the strong difference between the elastic and the inelastic channel. And there are intermediate, ambiguous ranges where directly scattered neutrons form an acceptable background in the inelastic channels. Part of this transitory range is due to the finite time it takes the chopper to open and to close. Other contributions include the following:

• In the present chopper, the graphite crystals are kept in place by aluminum corner pieces. When these pieces cross the primary neutron beam, they cause incoherent scattering into 4π, thereby sending a diffuse neutron cloud into the secondary spectrometer.

• Some directly scattered neutrons are retarded by phonon downscattering, causing a sampledependent extension of the chopper-closing phase.

• Multiple scattering (first scattering in the sample, second scattering by the sample environment or somewhere in the secondary spectrometer) also presents a sample-dependent contribution to the extended chopper-closing phase.

To parametrize the data acquisition daemon, the instrument responsible must make a relatively arbitrary choice which range of chopper phases shall be considered as indirect. In general, users of SPHERES are more interested in an excellent signal-to-noise ration than in maximum count rates. Therefore, a relatively strict chopperphase discrimination that favors signal-to-noise ratio on the expense of total count rates is preferred. For most measurements in the past years, we have set the indirect interval to 35^∘, which means that the duty cycle (the fraction of time actually used for incrementing scattering histograms) is no more than 35/120 = 0.29.

For the future, one might consider a small modification of the data acquisition code to simultaneously record energy histograms for different chopper-phase discrimination settings. Users could then decide ex post whether to analyze a “strict” data set with optimum signal-to-noise ratio, or a “loose” data set with higher count rates.

D. Resolution and Signal-to-Noise Ratio

The instrumental resolution can be empirically determined by measuring the elastic scattering of a solid. The scattering should be predominantly incoherent to guarantee sufficient intensity at all scattering angles. Vanadium, the elastic incoherent reference scattering par excellence, is not well suited because its absorption cross section at λ = 6.27 Åis more than three times larger than its scattering cross section. Therefore, we prefer a simple plastic foil (Hostaphan RN 250: a polyethylene terephthalate foil of 0.25 mm thickness) as a our reference scatterer.

In most experiments, however, the resolution is measured on the user sample, cooled to a low temperature where almost all scattering is elastic. This has the advantage that the sample geometry is exactly the same as in the production measurements.

Fig. 7 shows such a resolution spectrum from a user experiment. The Doppler drive was running with the maximum velocity amplitude of 4.7 m/s, resulting in a dynamic range of ±31 μeV. To determine the resolution width and the elastic amplitude, the centre of the elastic peak (|ℏω| < 0.6 μeV) has been fitted by a Gaussian,

FIG. 7: A resolution measurement with a user-provided sample (a powder of hydrogen-rich organic crystals at 3 K, in a flat cell calculated for 90% transmission). Cumulated data from the six large-angle detectors with the best signal-to-noise ratio. The solid line is a Gaussian fit to the central region |ℏω| < 0.6 μeV.

which is an excellent approximation though there are pronounced non-Gaussian wings at larger energy transfers. The full width at half maximum (fwhm) is obtained by multiplying the standard deviation of the Gaussian with √ 8 ln(2).

When the Doppler drive is run with small to moderate velocity amplitudes, the resolution at large scattering angles is typically between 0.62 and 0.65 μeV. At the maximum Doppler frequency used here, the resolution is slightly reduced (to 0.66 μeV), probably because of distortions of the monochromator support.

The signal-to-noise ratio (SNR) depends obviously on how we define the “noise”. If we average only over energies far off the central peak, say |ℏω| ≥ 20 μeV, then we obtain in the present example 1750:1. The SNR also depends on the sample: if the sample is too thin, then the sample-independent constant background contributes relatively more; if the sample is too thick, then many backscattered neutrons are lost when they retraverse the sample.

VII. COMPARATIVE STATUS AND PERSPECTIVE

In Table II, some design and performance parameters of the three backscattering spectrometers IN16, HFBS and SPHERES are compared. These parameters are intercorrelated as follows.

The instrumental resolution is within 5% proportional to the chosen thickness of the Si(111) monochromator/analyzer wafers. This confirms the fundamental importance of the crystal choice for the instrument performance, and it shows that all three instruments very closely attain the optimum resolution allowed by their crystal sets.

instrument IN16^a HFBS SPHERES reactor power (MW) 58 20 20 PG(002) speed (m/s) 75 250 100 Si(111) thickness (μm) 700 750 500 resolution fwhm (μeV) 0.8–0.9 0.93 0.62–0.65 flux at sample^b (10^6 s^{−1}) 1.2 2.4 1.8 rescaled flux^c 0.17 0.91 1 signal-to-noise ratio^d 1000^e 600 1750

aWith the unpolished, deformed monochromator/analyzer set that is used in almost all experiments.

bWhile chopper open. Au foil activation gives half this value. cFlux at sample, divided by resolution width and reactor power, arbitrarily normalized to the value of SPHERES dBest value reported for a non-absorbing sample of optimum thickness.

eNot mentioned in the instrument descriptions^{13–15,44}, but confirmed by instrument users.

TABLE II: Comparison of the three backscattering spectrometers IN16 of the ILL,^{13–15,44} HFBS at NIST,^{19} and SPHERES of the JCNS at FRM II.

Ceteris paribus, the flux at the sample should be proportional to the reactor power. In a very good approximation, it should also be proportional to the width of the energy band selected by the monochromator. Therefore, the figure of merit that describes the overall efficiency of the cold source, of the neutron guides, and of the primary spectrometer is flux divided by reactor power divided by resolution width. The comparison of this rescaled flux shows how much the compact design of HFBS and SPHERES, due to Schelten and Alefeld,^{17} is superior to the double-deflector layout of IN16.

The primary spectrometer transmission is also somewhat related to the signal-to-noise ratio: to maximize the latter, we installed in SPHERES ad hoc several slits that cut away some 10% of the incoming and of the backscattered beam in order to prevent neutrons from being transmitted through the closed chopper. The next chopper, currently under construction, will hopefully allow us to take away these slits and to increase the flux at the sample accordingly. Furthermore, in this new chopper the PG(002) deflector crystals will be moved with about optimum speed, which will result in yet another important gain in flux.

At small scattering angles, analyzers cannot be aligned to exact backscattering geometry because detectors must be placed outside the incoming neutron beam. This makes the resolution much wider and more assymetric than in the regular backscattering detectors. Supported by simulations, we have improved the resolution at small angles by breaking the azimuthal symmetry of the analyzer rings, as will be described separately. Acknowledgements

Building and commissioning SPHERES has been funded by the German Bundesministerium für Bildung und Forschung, projects 05NX8CJ1 and 03RI16JU1.

Foundations for SPHERES were laid by former project scientists Oliver Kirstein and Peter Rottländer, and by former project engineer Tadeusz Kozielewski. Many other colleagues from Forschungszentrum Jülich contributed to this project. We would like to acknowledge Ulrich Probst, Helga Straatmann and Thomas Koppitz for development of the chopper rotor, Ulrich Giesen and Ulrich Pabst for a vibrational analysis therof, Christoph Tiemann for his contribution to the Doppler drive, Gerd Schaffrath for the instrument housing, Manfred Bednarek for the electrical installations, Peter Stronciwilk and Marco Gödel for mechanical constructions, Vu Thanh Nguyen for the accurate glueing of the silicon crystals, Harald Kusche, Andreas Nebel, and Björn Poschen for technical support, and Alexander Ioffe for coordinating the JCNS outstation at FRM II. Sergej Manoshin and Alexander Ioffe contributed a simulation of the convergent neutron guide. Michaela Zamponi, the new instrument responsible since fall 2011, contributed to the last Au activation measurement.

We are grateful to Winfried Petry, Jürgen Neuhaus, and all the staff of FRM II for their hospitality. We thank Harald Türck for generous help at all levels of engineering, Christian Breunig for the neutron guide, Philipp Jüttner for the shutter construction, Helmut Zeising, Birgit Wierczinski and their entire team for smooth handling of the radioprotection necessities, and Ralf Lorenz for sensibly supervising our safety.

We thank our colleagues and friendly concurrents Bernhard Frick, Tilo Seydel, and Lambert van Eijck of the ILL for stimulating discussions and the open exchange of experiences and ideas. Victoria García Sakai, Goran Gasparovic, and Dan A. Neumann kindly showed us HFBS at NIST, and Andreas Meyer provided insights from his commissioning experience.

∗ Electronic address: j.wuttke@fz-juelich.de † permanent address: Grup de Caracterització de Materials, Departament de Física i Enginyeria Nuclear, Universitat Politècnica de Catalunya, Barcelona, Spain.

‡ deceased in 2008.

1 M. Bée, Quasielastic Neutron Scattering, Adam Hilger:

Bristol (1988).

2 R. Hempelmann, Quasielastic Neutron Scattering and

Solid State Diffusion, Clarendon Press: Oxford (2000). 3 H. Maier-Leibnitz, Nukleonik 8, 61 (1966).

4 B. Alefeld, M. Birr and A. Heidemann, Naturwiss. 56, 410 (1969).

5 B. Alefeld, Kerntechnik 14, 15 (1972).

6 G. Allen, R. E. Ghosh, A. Heidemann, J. Higgins and W. S.

Howells, Chem. Phys. Lett. 27, 308 (1974).

7 J. C. Cook, W. Petry, A. Heidemann and J. F. Barthélemy,

Nucl. Inst. Meth. A 312, 553 (1992).

8 O. G. Randl, H. Franz, T. Gerstendörfer, W. Petry, G. Vogl and A. Magerl, Physica B 234–236, 1064 (1997).

9 A. Heidemann, I. Anderson, M. Prager and W. Press, Z. Phys. B 51, 319 (1983).

10 T. Springer, Phil. Trans. Roy. Soc. Lond. B 290, 673 (1980).

11 B. Alefeld, T. Springer and A. Heidemann, Nucl. Sci. Eng.

110, 84 (1992).

12 A. J. Dianoux, Physica B 182, 389 (1992). 13 B. Frick, A. Magerl, Y. Blanc and R. Rebesco, Physica B 234–236, 1177 (1997).

14 B. Frick and M. Gonzalez, Physica B 301, 8 (2001). 15 B. Frick, Neutron News 13, 15 (2002).

16 N. de Souza et al., Emu project,

http://www.ansto.gov.au/research/bragg_institute/facilities/instruments/emu_-_high-resolution_backscattering_spectrometer.

17 J. Schelten and B. Alefeld, in Proceedings of the Workshop on Neutron Scattering Instrumentation for the SNQ, Maria Laach, 3-5 Sept 1984, edited by R. Scherm and H. Stiller (Berichte der Kernforschungsanlage Jülich, Jül- 1954), KFA: Jülich (1984).

18 P. M. Gehring and D. A. Neumann, Physica B 241–243, 64 (1998).

19 A. Meyer, R. M. Dimeo, P. M. Gehring and D. A. Neumann, Rev. Sci. Instr. 74, 2759 (2003).

20 M. A. Gonzalez and B. Frick, Appl. Phys. A 74s1, 474 (2002).

21 B. Frick, H. N. Bordallo, T. Seydel, J.-F. Barthélémy, M. Thomas, D. Bazzoli and H. Schober, Physica B 385– 386, 1101 (2006).

22 O. Kirstein, M. Prager, H. Grimm and D. Richter, J. Neutron Res. 8, 119 (1999).

23 O. Kirstein, M. Prager, T. Kozielewski and D. Richter, Physica B 283, 361 (2000).

24 O. Kirstein, T. Kozielewski, M. Prager and D. Richter, Physica B 291, 310 (2000).

25 O. Kirstein, T. Kozielewski and M. Prager, in Proceedings of the Deutsche Neutronenstreutagung 2001, edited by T. Brückel (Materie und Material, Vol. 8), FZ Jülich: Jülich (2001).

26 O. Kirstein, T. Kozielewski, M. Prager and D. Richter, Appl. Phys. A 74, S133 (2002).

27 P. Rottländer, T. Kozielewski, M. Prager and D. Richter, Physica B 350, e823 (2004).

28 SPHERES shares the guide NL6a with the monochromator

of MIRA 2 (3–6 Å). The alternate setup MIRA 1 uses an end position at the heavily curved guide NL6b (radius 84 m) to obtain very cold neutrons (8–30 Å).

29 Supplemental Material [URL...].

30 A. Meyer, J. Wuttke, W. Petry, A. Peker, R. Bormann,

G. Coddens, L. Kranich, O. G. Randl and H. Schober, Phys. Rev. B 53, 12107 (1996).

31 A. V. Skripov, J. C. Cook, D. S. Sibirtsev, C. Karmonik

and R. Hempelmann, J. Phys. Condens. Matter 10, 1787 (1998).

32 M. Hennig, B. Frick and T. Seydel, J. Appl. Cryst. 44, 467

(2011).

33 J. Wuttke, to be published.

34 U. Probst, H. Kämmerling, T. Kozielewski, H. Straatmann, P. Rottländer, T. Koppitz, European Patent EP 1706876 (2005).

35 A. D. Stoica and M. Popovici, J. Appl. Cryst. 22, 448

(1989).

36 J. Combet, B. Frick, O. Losserand, M. Gamnon and B. Gurard, Physica B 283, 380 (2000).

37 A. Gohr et al., Dokuwiki, http://www.dokuwiki.org/

(2004–2011).

38 M. Drochner, H. Kleines, M. Kirsch, T. Kozielewski,

M. Prager and P. Rottländer, in 14th IEE-NPSS Real Time Conference: Stockholm (2005).

39 J. Wuttke, lmfit - a C/C++ routine for Levenberg-

Marquardt minimization with wrapper for least-squares curve fitting, based on work by B. S. Garbow, K. E. Hillstrom, J. J. Moré, and S. Moshier,

http://joachimwuttke.de/lmfit.

40 O. Ben-Kiki, C. Evans and I. döt Net, YAML Ain’t Markup

Language, http://www.yaml.org.

41 H. Kleines, F. Suxdorf, M. Drochner, P. Rottländer,

M. Prager, T. Kozielewski and A. Budwig, KEK proceedings 2005-18, WEP25 (2005).

42 J. Wuttke, Slaw: neutron scattering

histograms to scattering law converter,

http://apps.jcns.fz-juelich.de/slaw.

43 J. Wuttke, absco — absorption corrections for X-ray and neutron diffraction,

http://apps.jcns.fz-juelich.de/absco.

44 IN16 characteristics, http://www.ill.eu/instruments-support/instruments-groups/instruments/in16/characteristics/.