arXiv:1009.2262v1 [physics.ins-det] 12 Sep 2010

Preprint typeset in JINST style - HYPER VERSION

A hardware implementation of Region-of-Interest selection in LAr-TPC for data reduction and triggering

B. Baibussinov^a, S. Centro^{a∗}, K. Cieslik^{a,b}, D. Dequal^a, C. Farnese^a, A. Fava^a, D. Gibin^a, A. Guglielmi^a, G. Meng^a, F. Pietropaolo^a, C. Rubbia^{c,d}, E. Scantamburlo^a, F. Varanini^a, S. Ventura^a a Istituto Nazionale di Fisica Nucleare and Dept. of Physics “G. Galilei", Via Marzolo 8 35131 Padova, Italy b On leave form Institut Fizyki Jadrowej PAN, Krakov, Poland c Laboratori Nazionali del Gran Sasso dell’INFN, I-67010 Assergi (AQ), Italy d CERN, European Laboratory for Particle Physics CH-1211 Geneve 23, Switzerland E-mail: sandro.centro@pd.infn.it

ABSTRACT: Large Liquid Argon TPC detectors in the range of multikton mass for neutrino and astroparticle physics require the extraction and treatment of signals from some 10^5 wires. In order to enlarge the throughtput of the DAQ system an on-line lossless data compression has been realized reducing almost a factor 4 the data flow. Moreover a trigger system based on a new efficient on-line identification algorithm of wire hits was studied, implemented on the actual ICARUS digital readout boards and fully tested on the ICARINO LAr-TPC facility operated at LNL INFN Laboratory with cosmic-rays. Capability to trigger isolated low energy events down to 1 MeV visible energy was also demonstrated.

To be submitted to JINST

KEYWORDS: Liquid Argon; TPC; neutrino.

∗Corresponding author

Contents

1. Introduction 2

2. Cosmic-ray test-run with ICARINO test-facility at LNL 2

3. The new hit finding algorithm: a trigger from the wire signals 8 3.1 Single hit detection in Collection view 12 3.2 Study of the GTO signal in Collection view 12 3.3 Internal trigger based on the Collection GTO signal to search for tracks in Induction view 13 3.4 A low energy event trigger based on GTO signals 15

4. Conclusions 17 Bibliography19

– 1 –

1. Introduction

Developments in neutrino and astroparticle physics will require the realization of gigantic detectors with mass of order of hundreds kton or megaton to be installed in underground laboratories. Such an apparatus will be addressed to study long base-line neutrino oscillations at accelerators as well as from cosmic sources including Supernovae, and to test the barion matter stability searching for proton decay. Among the proposed detection techniques, the Liquid Argon TPC detector (LAr- TPC) is expected to play a special role, accounting for the excellent event imaging capability and the superior performance concerning event detection efficiency and identification which allows to keep the detector size in the multikton mass range with respect to the megaton masses required for example by the water Cherenkov technique. Thus the efforts linked to the underground site excavation and infrastructures, as well as the engineering needs, can be largely reduced. The successful starting operations of ICARUS-T600 [1] 600 ton LAr-TPC at INFN LNGS underground laboratory using the CERN CNGS neutrino beam, represent a milestone towards the realization of large LAr-TPC mass detectors for neutrino and astroparticle physics.

The extrapolation to multikton mass LAr-TPC detector requires some improvements in different aspects of the technology, such as cryogenic plant, LAr purity and detector read-out [2]. Moreover extraction and handling of the corresponding huge amount of data will constitute a not negligible aspect to be addressed in the detector exploitation. In the present ICARUS-T600 about 54000 channels of the wire TPC are read-out. To increase the bandwidth of the DAQ system two improvements have been studied for the TPC wire signal read-out: an on-line lossless factor 4 data compression, and a new algorithm to detect the local ROIs (Region of Interest) of each event avoiding the full acquisition of the detector. In particular the second one will allow triggering isolated low energy events down to 1 MeV visible energy using directly the wire signals.

In the following the algorithm description and implementation on the TPC read-out board is described as well as the performance, tested with the ICARINO LAr-TPC facility in the framework of the ICARUS R&D activity at the INFN-LNL laboratories in Legnaro (Italy).

2. Cosmic-ray test-run with ICARINO test-facility at LNL

A detailed description of the ICARINO experimental set-up (Fig. 1) can be found elsewhere [3]. The TPC consists of two vertical electrode planes, 32.6 × 32.6 cm^2, acting as anode and cathode, laterally delimited by 4 vetronite 29.4 × 29.4 cm^2 boards supporting the field-shaping electrodes (30 strips of gold-plated copper) resulting in a 38 kg LAr active mass (Fig. 2). In order to facilitate the LAr re-circulation in the TPC active volume, the cathode at the front of the chamber has been realized with a thin etched stainless steel grid.

The chamber is contained in a stainless steel cylindrical vessel, whose upper face is an ultra high vacuum flange hosting the feed-throughs for vacuum, LAr filling and re-circulation, high voltage cables and read-out electronics. The whole detector vessel is contained in an open-air stainless steel dewar, which is initially filled with commercial LAr acting as cryogenic bath for the ultrapure LAr injected in the detector vessel.

The read-out anode’s electrodes facing the drift volume are two parallel 96 stainless steel wire planes spaced by 3 mm and with 3 mm pitch: the first one works in induction mode and is made of

– 2 –

Figure 1. The ICARINO test-facility set-up in LNL.

Figure 2. Images of LAr-TPC at LNL: anode (left) and cathode (right) structure of the TPC.

vertical wires while the second one collects the drifting electrons and is made of horizontal wires. A third wire plane electrically biased, called grid, is inserted 3.5 mm in front of the Induction wires. It acts as an electromagnetic shield improving the induction signal sharpness and reducing

– 3 – the noise due to the HV biasing. Finally, another thin stainless steel grid has been inserted behind the Collection wire plane and put to ground to confine the electric field around the wires.

A -14.8 kV voltage is applied on the cathode corresponding to a uniform electric field of 474 V/cm in the drift volume. In order to ensure a full transparency of the Grid and of the Induction plane to drifting electrons, the potential is fixed at 350 V for the Collection wires, - 100 V for the Induction wires and - 350 V for the Grid. This set of values was chosen after a careful study on the reconstructed muon tracks collected in several preliminary test runs, optimizing the electric field uniformity in the whole drift region.

The TPC vessel is filled with ultra pure LAr, following the procedure described in [3]. First the chamber is evacuated to at least 10^{−3} Pa with turbo-molecular pumping system for several days, to maximize degassing of the detector materials. Then the chamber is cooled down in the LAr bath as fast as possible and is filled with commercial LAr purified by the Oxysorb/Hydrosorb filter. An additional small evacuated tank immersed into a LAr bath, acting as a cryogenic trap, is connected to the TPC vessel during the first phase of the filling to remove the gaseous Argon, produced by the evaporation on the warm inner detector surfaces and possibly contaminated by degassing. This additional vessel is disconnected soon after the completion of the filling procedure.

The level of the LAr cryogenic bath, in which the detector vessel is immersed, is lowered with respect of that in the inner detector, to increase the heat losses and allow the recirculation system to start. The gaseous Argon (GAr), evaporating from the liquid in the detector vessel, is purified by the Oxysorb/Hydrosorb filter, condensed in a passive heat exchanger (LAr bath) and finally reinjected at the bottom of the detector. The total LAr volume is completely recirculated every 2.5 days.

The read-out electronics is a subset of the one implemented in the ICARUS-T600 detector [1]. It’s designed to provide continuous digitization and waveform recording of the signals from each TPC wire; the chain is composed of three basic units, each one serving 32 channels: the Decoupling Board, the Analog Board, and the Digital Board.

The Decoupling Board receives 32 analogue signals from the chamber and passes them to the analogue board via a decoupling capacitors; it also provides wire biasing voltage and the distribution of the test signals.

The Analogue Board hosts the front-end amplifiers and performs 16:1 channel multiplexing and 10-bit digitization at 40 MHz rate (i.e. 400 ns sampling period per channel, corresponding to ∼ 0.6 mm drift distance). The overall gain is around 5-6 ADC counts/fC, i.e. about 1000 electrons per count, thus setting a minimum ionizing particle signal (∼ 1 fC/3mm) at about 10 ADC counts with a dynamic range of about 100 mip’s. To match the different signal’s shapes two versions of this board have been used, differing for the integration time constant. The Collection version has a time constant of 3 μs, much shorter than the signal duration, and is used for the unipolar signals coming from Collection wires; the area of the shaped signal is proportional to the charge. The Induction version has a time constant of 100 μs, much longer than the signal duration, and is used for the bipolar current signals coming from the Induction wires. In the ICARUS detector the Collection version is also used to instrument the Induction plane facing the drift volume, not instrumented in the present test-facility.

– 4 –

Figure 3. Block diagram of the LAR-TPC electronic read-out.

The Digital Board (Arianna) acts as a 32 channels, 10 bit wide, waveform recorder. It continuously reads the data out of the V791 Board, stores them in multi event circular buffers (MEBs) and analyzes them according to a complex, programmable logic. MEBs length can be chosen among 7 different values, ranging from 64 to 4096 t-samples (1 t-sample = 400 ns), corresponding to a drift distance from 38 mm to 2.5 m at nominal electric field; this feature permits a segmentation of data along the drift direction, defining Regions of Interest (ROIs). As soon as a trigger signal is received, the active buffer is frozen, writing operations are moved to the next free buffer, and the stored data are read out by the DAQ.

Once the read out is completed, the frozen buffer is released and it’s available to be overwritten by new incoming data. This configuration guarantees no dead time, at least until the maximum DAQ throughput (up to 1 Hz, i.e. 1 full-drift event per wire per second in the ICARUS-T600 configuration) is reached.

The buffer data are formatted in four different modes:

• RAW-DATA: no compression is performed, and data (Daedalus output + wire signal waveform) are written to the MEB as they are (Fig. 5);

– 5 –

Figure 4. Picture of the digital board equipped with the new SuperDaedalus chip, implementing the new hit-finding algorithm.

Figure 5. Raw data model of a 16 bit word.

• COMPRESSION 4: for each set of 4 wires, the signal amplitude value for a t-sample is compared to the previous one. If the difference is less than or equal to ± 7 ADC counts the four 4 bit differences are stored in a 16 bit word (Fig. 6). If one or more differences are above that value (OVERFLOW) all the four differences are stored each one in a 16 bit word, flagged setting to “1000” the 4 more significant bits (“1000 = 8”, out of the compression 4 difference range);

• FULL-DIFFERENCE: for each wire the difference between two consecutive t-samples is written using 10 bit. Since the difference is preceded by the string "100000" this mode is equivalent to compression 4 during an overflow (Fig. 6, bottom);

• COMPRESSION 2: for each wire the difference is always written using 8 bit. For “physical” signals it never overflows (Fig. 7).

– 6 –

Figure 6. Compression 4 (top) and overflow (bottom) data model of a 16 bit word.

Figure 7. Compression 2 data model of a 16 bit word.

In addition in the ICARINO test-facility set-up both Induction and Collection boards are equipped with a new chip (SuperDaedalus) which implements a new hit-finding algorithm, here described in the following (Fig. 4).

The test-run described here addressed the study of the performance of the new digital boards with trough-going cosmic ray muons both in terms of data compression, and ROI definition capability. The system was run with a global stop signal provided by a trigger system based on external scintillation counters or on the internal TPC wires signal processed by the new SuperDaedalus chip that produces a Global Trigger Output (GTO) from each digital board.

All the data compression modes have been extensively tested in several runs. Thousands of test pulse and single cosmic muon events, both in raw data and compression 4 modes, were collected and analyzed to test the functionality of the compression algorithm. No difference was found between the two modes by comparing the pulse-height and the difference between contiguous tsamples (Fig.8).

About 30% of the events with single muon tracks resulted to be affected by overflows (Fig. 9). The distribution of the number of overflows per event (Fig. 10) shows on average 1 overflow per event. The detailed study of event reconstruction and signal shape in compression 4 mode next to the overflows shows no anomaly, no signal truncation or distortion.

The distributions of hits number and amplitude per track confirm that no difference is present between compression 4 mode and raw data (Fig.11), allowing to validate the new firmware of the digital boards. Therefore an effective reduction factor 3.9, with a consequent reduction in the read out time of the same factor, was established.

– 7 –

Figure 8. Distribution of the wire signal pulse-height (left) and of the corresponding differences between two consecutive t-samples (right) for both compression 4 (black) and raw (red) data modes (empty events).

3. The new hit finding algorithm: a trigger from the wire signals

As observed during the 2001 test-run with a half of ICARUS T600 detector the hit signals on the Collection wires produced by a through-going muon are characterized by a pulse-height in the range 10÷30 ADC counts depending on the track inclination and extending over ∼ 25 t-samples. Data exhibit also the presence of a ∼ 4 ADC counts peak to peak high frequency noise with period less than 10 t-samples overlapped to a 10 ADC counts low frequency oscillation of the baseline with a ∼ 1500 t-samples period.

In order to extract the hit signal from the wire chambers, a dedicated algorithm had been developed with the aim of filtering out on-line at the same time both the low and high frequency components of the noise by means of a double rebinning technique applied to each wire signal. Since quick response time would be essential for trigger purposes, the algorithm was conceived as simple as possible to be easily implemented in a hardware frame. It consists in calculating the average of the signal amplitude Q(t) over a short ∼ 10 t-samples and a long ∼ 250 t-samples time interval, to treat the high and the low frequency components respectively:

Q_{short} (t_j;t_{j+10}) =

j+10

∑_{i= j}^{Q(t}i^{)/10 Q}long^{(t} j^{;t} j+250^{) = j+250}∑_{i= j}^{Q(t}i^{)/250. (3.1)}

where the binning interval sizes have been initially chosen according to hit signal features compared to the noise; the presence of the physical hit is then recognized requiring the difference S(t) to exceed a fixed threshold Q_{thr}:

S(t) = Q_{short} (t) − Q_{long}(t) ≥ Q_{thr}. (3.2)

The performance of this algorithm as well as the parameter choice was studied through a software analysis of the cosmic muon data collected during the 2001 test run. As a initial result, the algo-

– 8 –

Figure 9. Event with 1 (top) and 2(bottom) overflows correctly reconstructed within the Compression Mode 4.

rithm turned out full efficient (∼ 100%) in the physical hit recognition with a negligible fake rate (≤ 3%) for threshold values Q_{thr} = 4 ADC counts over a wide range of the long rebinning interval (100 ÷ 300 t-sample).

– 9 –

Figure 10. The distribution of the number of overflows per event (Compression Mode 4).

A slight modification with respect to the software algorithm was introduced in the hardware implementation of this simple algorithm. The interval into which the signal averages (long and short) are calculated is a “sliding window” over the full data recording (DR-slw). Moreover new values of the binning interval sizes, more suitable for the hardware implementation, were chosen: 8 and 128 t-samples for the short and long binning respectively. At each clock cycle the average over 8 and 128 t-sample is recalculated adding the value of the sample t0 and subtracting that of sample t0-8 and t0-128 respectively. When the difference of short and long rebinned signals is above a threshold for at least 3 t-sample, a PEAK signal is output (fig. 13).

Possible impact of these modifications on the algorithm performance has been investigated on a sample of real muon events collected in Pavia. The result confirms the possibility to obtain full detection efficiency (∼ 99%) together with a negligible fake signal rate (∼ 1%) applying a threshold value Q_{thr} = 6 ADC counts, slightly higher than for the original algorithm (fig. 14).

The hit identification procedure can also be used for the recognition of a ROI and the generation of an internal trigger. The block diagram of the on-line algorithm implementation is shown in Fig. 15. Serial data (10 bit for each channel) are de-multiplexed to enter the rebinning algorithm. Through the parameter "polarity" it’s possible to set the trigger on the rising or on the falling edge. In particular the falling edge is used for the Induction wires, due to the presence of an undershoot on those signals.

To get a trigger from the PEAK signals a majority stage has been included. Dividing the 32 wires in groups of 16, and taking the logical OR of the two majority, the rate of fake triggers can be reduced, while keeping the system sensitive to the short tracks (16 wires correspond to 5 cm). A possible drawback of this solution could arise for inclined tracks: as the PEAK signals may not overlap, the efficiency of the majority would decrease with the angle (Fig. 14) . To avoid this

– 10 –

Figure 11. Distribution of the hit number per track (top) and of their corresponding amplitude in ADC counts (bottom) in Compression Mode 4 and Raw data (muon events).

situation, the final stage of the SuperDaedalus performs a stretching of the PEAK signals among four values, 25, 25, 75, and 125 μs; analysis made on data of the 2001 test-run showed that the efficiency never fall below 99% for traces of all angles if a stretchng of 50 or 75 μ s is applied.

In order to determine the optimal parameters for the new Collection boards, cosmic muons track events have been collected with the ICARINO test-facility with the external scintillation counter trigger. Then the new GTO internal signal from Collection boards has been used to trigger cosmic muons and to study the performance in Induction view also.

Finally a particular configuration of internal TPC trigger, aiming to the detection of isolated low energy events (solar neutrino like) has been set-up. This last measurement represent a crucial

– 11 –

Figure 12. Typical signal shape of a Collection wire. Both the high and low frequency components of the electronic noise can be clearly seen.

test in view of the future development of huge LAr-TPC detectors, since the localization of low energy events represents a challenging item for large volume detectors.

3.1 Single hit detection in Collection view

The hit finding efficiency on the single wire in Collection view has been measured as a function of the threshold value Q_{thr} ranging in the interval 4÷ 10 # ADC, with the stretching parameter set at 50 and 75 μs and requiring the coincidence within 10 μs with the hit identified in the off-line analysis. Among all the vertical and 45^∘ inclined muon tracks, collected by means of the external trigger, only the through going muons without energetic δ rays were selected (fig. 16). The statistic was ∼ 1000 and ∼ 500 events for vertical and inclined tracks respectively.

As a result an almost full single hit finding efficiency was measured for different track inclinations at low Q_{thr} which degrades with increasing the Q_{thr} threshold. Moreover efficiency values above 95 % wwere measured for Q_{thr} ≤ 7 ADC counts independently of the dip angle (fig. 17). A dedicated run with random external trigger was performed to measure the corresponding fake hit detection rate on the single wire recognizing the presence of the PEAK signals on the whole 1024 tsamples interval in empty events (i.e. with no hits identified by the off-line analysis software). The rate of fakes per event was rapidely decreasing down to 10^{−3} with threshold for both Collection and Induction views for Q_{thr} ≥ 6# ADC (fig.18) (1000 events overall collected statistics). The obtained results demostrated the single hit finding reliability, allowing to work satisfactory, with high efficiency and low associated fake hit signals.

3.2 Study of the GTO signal in Collection view

Characteristics and performance of the GTO signal in Collection view were studied on a sample of

– 12 –

Figure 13. Example of the DR-slw algorithm applied to the wire signal in previous figure: Q_{short} (t) (top), Q_{long}(t) (middle) and S(t) (bottom).

clean through-going muons tracks, selected among vertical and 45^∘ cosmic muon events collected with the external trigger system (fig. 19). GTO signals from two out of the three Collection boards were recorded for a fixed threshold value Q_{thr} = 6 (chosen relying on the analysis described in the previous section), for different values of the majority (M = 8, 12,15) and stretching parameters.

Time delay of the GTO signal with respect to the first track hit, depends mainly on the track dip angle due to time displacement of the PEAK signals on consecutive wires and on the required majority level. Therefore the GTO signal efficiency was determined into a suitable time window of 50 and 300 t-samples depending of the track dip. An almost 100 % efficiency has been observed for majority values up to M = 15 for inclined and vertical tracks for both boards (fig. 20) even with the shortest stretching parameter. Actually small inefficiencies for in the first board for the inclined tracks sample are caused by events not completely contained, entering the TPC volume after the 24^{th} wire and then with less than 8 hits on the first board.

3.3 Internal trigger based on the Collection GTO signal to search for tracks in Induction view

An internal trigger system based on the coincidence of the GTO signals from the first and the

– 13 –

Figure 14. Hit identification efficiency for three different track inclination (top) and fake frequency (bottom) of the DR-slw algorithm as a function of the discrimination threshold in data collected in Pavia with the a semimodule of T600.

third boards in Collection view was set-up. A sample of events of through going μ almost parallel to the TPC wire planes were collected and used to study the performance of the peak signal in Induction view. In particular, only tracks with 45^∘ ± 5^∘ inclination with respect to the Induction and Collection wire orientation were selected (fig. 21).

The hit finding efficiency on the single wire was ≥ 97% up to threshold values Q_{thr} = 6# ADC, like in Collection view, but decreased more quickly with Q_{thr}.

– 14 –

Figure 15. Block diagram of the DR-sw hit finding procedure and generation of a local trigger signal.

In order to avoid any problem concerning unconfined muon tracks, the efficiency of the GTO signal from the central board in Induction view was measured with an upgraded internal trigger were by coincidence of the GTO signals from the the first and the third Induction boards was included.

The obtained result, ∼ 98.0 ± 0.3% efficiency (Q_{thr} = 5# ADC, M = 6 and stretching 50 ÷ 75 μs) , is really satisfactory also because almost half of the small inefficiency was due to vanishing tracks with few hits above threshold (fig. 22). as recognized by a visual scan of the events.

3.4 A low energy event trigger based on GTO signals

A challenging item for huge mass detectors is the selection of localized low energy events, such as for example that ones involved in the search for solar or Supernovae neutrinos interactions. Thus an interesting application of the internal trigger based on TPC wire signals was realized in Icarino vetoing the GTO signal from the central board with the coincidence of the GTO signals from the two lateral boards. The collected events, mainly γ rays and neutrons (Figs. 23, 24 and 25), show on average a hit multiplicity per event in Collection(Induction) view n_h = 7.3(7.9), for Q_{thr} = 5# ADC and M = 3, and n_h = 9.1(14.6), for Q_{thr} = 6# ADC and M = 4, on an overall statistics of 4857 and 21483 events respectively.

The deposited ionization energy E_{dep} was measured event by event from the integral of the wire signal recorded in Collection view A_i, according to E_{dep} = Σ_iA_i^C_e ·^{Eion}_R , where e = 1.6 · 10^{−4} fC is the electron charge and E_{ion} = 23.6 eV is the ionization energy in LAr. C = (1.39 ± 1%) · 10^{−2}

– 15 –

Figure 16. Visualization of an inclined muon track in the wire/t-sample plane. The hit signals in the Collection (left) and Induction (right) view are drawn as a function of the drift coordinate; the red line represents the peak signal for the threshold value Q_{thr}= 6 # ADC.

fC/(# ADC · t-sample) is the on electronic chain calibration constant, extracted in a dedicated testpulse run. Corrections to E_{dep} from electron recombination was included (the attenuation due to Argon impurities is negligible because of the short drift path). The resulting energy spectra of the events collected in the two runs extended up to 100 MeV (Fig. 26), with an average value of 6.9

– 16 –

Figure 17. Mean hit identification efficiency on the single wire in Collection view as a function of the threshold value Q_{thr} measured for vertical and 45^∘ inclined muon tracks. As a reference the values obtained with the software simulation of the DR-slw algorithm are also represented.

and 15.6 MeV respectively; the most energetic events referred to inclined muon tracks with large δ -rays traversing only the central region of the TPC.

4. Conclusions

The realization of a multikton mass LAr-TPC detector requires the extraction and the treatment of the corresponding huge quantity of data from some 10^5 channels. To increase the troughtput of the DAQ system an on-line lossless factor ∼ 3.9 data compression based on the recording of the difference between the wire signal at two consecutive time sample was realized and tested with cosmic rays at the INFN LNL ICARINO test-facility.

A new algorithm based on a double rebinning - sliding window technique was studied and implemented on the read-out board for a full efficiency detection of a single hit signal on a TPC wire. In such a way a local trigger system was successfully implemented and tested with cosmic rays allowing triggering isolated low energy events down to 1 MeV visible energy using directly

– 17 –

Figure 18. Average fake trigger rate per event on the single wire in Collection and Induction view, measured on the whole 1024 t-samples interval.

the wire signals. Moreover it’s allow to define local ROIs (Region of Interest) avoiding a full acquisition of the detector when the event is localized.

Acknowledgments

The work described in this paper has been carried out in the framework of the PRIN 2007 (Progetto di ricerca scientifica di Rilevante Interesse Nazionale) funded by the MIUR (Ministero dell’Istruzione, dell’Universita’ e della Ricerca). The project, whose title was "Techniques for signal detection and treatment in large Liquid Argon apparatuses for new generation astro-particle physics experiments", involved the Universities and INFN section of L’Aquila, Padova, Pavia.

– 18 –

Figure 19. Visualization of a vertical through-going muon track in the wire/t-sample plane, i.e. parallel to the TPC wire planes, both in Collection and Induction views. The red contour define the region in which a peak signal is present for a threshold value Q_{thr} = 6# ADC, while the blue line indicates the GTO signal for a Majority value M = 8. The profile of a typical GTO signal recorded in Collection view is also shown (bottom).

References

[1] S. Amerio et al.,ICARUS Coll., Nucl. Instr. and Meth. A527 (2004) 329. [2] D. Angeli et al., JInst 4:P02003 (2009). [3] B. Baibussinov et al. JInst 5:P03005 (2010).

[4] D. Gibin et al., A study of wire signal discrimination, ICARUS-TM/04-10, 2004.

D. Gibin et al., An improved study of wire signal discrimination, ICARUS-TM/05-04, 2005. [5] S. Amerio, Consideration on the ICARUS read out and on data compression, ICARUS-TM/2002-05, 2002.

[6] G. Meng, Modification of V789 digital board,2005.

[7] G. Meng et al., Improvements of the ICARUS DAQ system, ICARUS-TM/2008-02, 2008. [8] B. Baibussinov et al., A real time algorithm for track finding in ICARUS experiment, ICARUS-TM/2008-04, 2008.

– 19 –

Figure 20. Efficiency of the GTO signal per event in Collection view for the smallest value of the stretching parameter, 25 ÷ 50μs, and M = 8, 12, 15, for vertical and 45^0 tracks as measured in the second and third board.

– 20 –

Figure 21. Visualization of a through-going muon track parallel to the TPC wire planes and with 45^∘ ± 5^∘ inclination with respect to the Induction and Collection wires orientation. The values of the threshold and majority parameters are Q_{thr} = 6# ADC and M = 8 respectively.

– 21 –

Figure 22. Example of GTO signal inefficiency of the central Induction board for an vanishing muon track recorded with internal trigger. On the right it can be noticed that the number of hits identified by the DRslw algorithm (red contour), even if larger than that one identified via software (blue stars), isn’t enough to generate a GTO signal, which is instead incorrectly generated by correlated noise (blue contour).

– 22 –

Figure 23. Isolated low energy event recorded in Icarino with the solar neutrino like internal trigger, for Q_{thr} = 6# ADC and M = 4, visualized in the Collection (top) and Induction (bottom) views. On the right a zoomed view of the region interested by the event is shown.

– 23 –

Figure 24. Isolated low energy event recorded in Icarino with the solar neutrino like internal trigger, for Q_{thr} = 6# ADC and M = 4, visualized in the Collection (top) and Induction (bottom) views. On the right a zoomed view of the region interested by the event is shown.

– 24 –

Figure 25. Isolated low energy event recorded in Icarino with the solar neutrino like internal trigger, for Q_{thr} = 5# ADC and M = 3, visualized in the Collection (top) and Induction (bottom) views. On the right a zoomed view of the region interested by the event is shown.

– 25 –

Figure 26. Deposited energy spectrum in Collection view E_{dep} (left) and corresponding n_{hits}, E_{dep} distribution of events recorded with the solar neutrino like internal trigger, for Q_{thr} = 6# ADC and M = 4 (top) and Q_{thr} = 5# ADC and M = 3 (bottom).

– 26 –